ACUSON P500 2.0 Configuration
A Step by Step document to guide the learner through configuring the ACUSON P500 system.
SIEMENS Healthineers System Configuration ACUSON P500™ Ultrasound System, Release 2.0 Product and System Training HC US MSC S SET Page 1 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 This presentation provide a review of the configuration menu for the ACUSON P500 2.0 system. Objectives SIEMENS Healthineers Access the Configuration Menu Identifying the Configuration Menu Choices Making Changes to the Configuration Menu Review Configuration Menu HC US MSC S SET Page 2 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Locate the Configuration Menu SIEMENS Healthineers Patient Exam Review Con New Report Browser Depth/Zoom Config Movert Help Cip Ster UO 1 UD 2 Freeze Shaver HC US MSC S SET Page 3 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: The Configuration Menu can be accessed from the Control Panel. Pressing the key will access the set-up menu. Objectives SIEMENS Healthineers Access the Configuration Menu Identifying the Configuration Menu Choices Making Changes to the Configuration Menu Review Configuration Menu HC US MSC S SET Page 4 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 The Configuration Menu SIEMENS Healthineers Configuration Custom Keys General 1 General 1 General 2 Depth and Zoom Knob Direction General 2 Custom Keys Increase Depth · Clockwise Counter-Clockwise Custom Keys Display Display Exam Increase Zoom Clockwise . Counter-Clockwise Soft Keys Exam M&R Soft Keys Report Key Function ............ Text Annotation M&R Body Marker Print/Store 1 Image Store Report Wireless Print/Store 2 Network Image Store Text Annotation Servic Clip Store Clip Store Body Marker Archive Left Pedal Freeze Wireless Authorization Right Pedal Image Store Network Printer Barcode User Defined 1 None Service Utility User Defined 2 Arrow Clip Store Stress Echo Clip Store Archive eSieScan Biopsy TEQ Update Authorization Configure New Key << Steer Printer Not Used Steer >> Barcode Text A Text B Utility DVR Stress Echo Needle Visualization Pointer eSieScan None ave Cancel HC US MSC S SET Page 5 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 The Configuration Menu is broken down in to 21 sections, but many of the choices are defaulted for the standard use. If you are familiar with ACUSON X Family or NX Series configuration menus, it has a similar look and operation. Stress echo and eSiescan fields were added with 2.0 as well as other changes in the overall menu. Objectives SIEMENS Healthineers Access the Configuration Menu Identifying the Configuration Menu Choices Making Changes to the Configuration Menu Review Configuration Menu HC US MSC S SET Page 6 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 System Specifications SIEMENS Healthineers General 1 Use Touchpad to place cursor Organization Hospital Name ACUSON * · Type in information Regional Settings Language English Date 3/5/2015 Date Format Day/Month/Year Time Drop-down menus 11:27: 9 + Height and Weight Units . Metric . U.S. Time Zone (UTC-08:00) Pacific Time (US & Cv Audio Power Off · Scroll slide bar  Beep on Button Press * Shut Down with Confirmation Dialog V Beep after Store Completes Shut Down Shut Down with Confirmation Dialog Select the setting Beep Volume Doppler Volume . 50 Touchpad Behavior Press SAVE to make changes Touchpad Travel Speed Medium Set By Tap · On off Save Cancel HC US MSC S SET Page 7 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Making Changes: Within each section, use the touchpad to place the cursor and then • Type in the information using the keyboard • Activate the drop down menu and make the selection • Adjust the scroll bar for the desired number • Select the desired setting. • Always press the “Save” button before exiting the menu. Objectives SIEMENS Healthineers . Access the Configuration Menu Identifying the Configuration Menu Choices Making Changes to the Configuration Menu Review Configuration Menu HC US MSC S SET Page 8 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 General 1 SIEMENS Healthineers . Configuration General 1 General 1 General 2 Organization Custom Keys Hospital Name ACUSON Display Exam Soft Keys Regional Settings General 1 M&R Report anguage English Date 2/23/2016 Text Annotation Date Format Day/Month/Year Time 5 :23:58 Body Marker General 2 Wireless Height and Weight Units . Metric OU.S. Time Zone (UTC-08:00) Pacific Time (US & Cv Network Service Custom Keys Clip Store Audio Image Settings Archive Authorization Beep on Button Press Image Presets Standard Display * Printer V Beep after Store Completes Standard Barcode Americas Utility Beep Volume Power Off Europe Stress Echo Asia eSieScan Doppler Volume - 50 Shut Down with Co MEA (Middle East Africa) Touchpad Behavior Touchpad Travel Speed Medium Set By Tap · On off Save Cancel HC US MSC S SET Page 9 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: The First key is “General 1”. This is where the hospital name is added so that it appears on the patient banner. Access regional presets in this menu. Once the preset selection is made, the system will reboot. It is important for the first power on, to confirm the date and time zone is correct. Selecting Shut down, with bypass a confirmation dialog and may be a choice for the Emergency Medicine department, where speed is very important. The touchpad speed defaults to medium but can be changed if the user needs a different speed. General 2 SIEMENS Healthineers Configuration General 2 General 1 Patient ID General 2 Custom Keys Save an image of the Patient screen after registering a patient. Display General 1 Exam Hide Patient Banner Soft Keys M&R General 2 Report Common Mode Text Annotation Body Marker Delete Body Marker on Unfreeze Custom Keys Wireless Delete Text on Unfreeze Network Service Body Marker Location Upper Right · Lower Right Display Clip Store Archive Arrow Size Small (Type2) Authorization Patient Registration Long Form Printer Barcode Utility Transducer Power Stress Echo eSieScan Reduce Transducer Power in bare status Off Off 5 secs 1 min 3 mins 5 mins 10 mins Save Cancel HC US MSC S SET Page 10 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: The next screen is General 2. Under the Patient ID, checking the save and image of the patient screen after registration , automatically stores this screen as part of the exam. The default is “not-checked”. Select the body marker location. The new transducer Power Saver is activated here. The default is Off. Setting a selection will save transducer life by reducing Power to 1-10%when not in use. This will be seen in the data field. Custom Keys SIEMENS Healthineers Custom Keys Depth and Zoom Knob Direction Increase Depth . Clockwise Counter-Clockwise * Increase Zoom Clockwise · Counter-Clockwise General 1 General 2 Key Function Print/Store 1 Image Store Custom Keys Print/Store 2 Image Store Clip Store Clip Store Display Left Pedal Right Pedal Image Store Exam User Defined 1 None Soft Keys User Defined 2 None Configure New Key Not Used · Close Study * Save Cancel HC US MSC S SET Page 11 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Under Custom keys, the Depth and Zoom keys can be programmed for clock-wise or counterclockwise depending on the user preference. The defaults are shown here. The Store/Print keys, Clip Store , Foot Pedal and the two user defined keys can be set to the customer preferences. Using the touchpad, place the cursor and select using the drop- down menu to make the selection. The new Key (located on the keyboard) will automatically close the previous study to allow for immediate scanning without patient info added. When selected in this menu. This is helpful for Emergency Room scanning when they primarily want to do a Fast exam or just use to look at an area quickly. Custom Keys SIEMENS Healthineers Print/Store Clip Store Foot Pedal User-defined PC Printer Image Store Freeze Full D Image Store Clip Store ·Image Store Full M Clip Store ·Clip Store Arrow Image Store and PC ·Caliper Clip Store Printer ·Update CW* ·TEQ Update Pointer* .U/D Flip << Steer /Steer >> ·L/R Flip Text A / Text B HC US MSC S SET Page 12 | Unrestricted @ Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: These are the drop down choices for the keys in the Custom Key menu. One Note: Set-up the Steer (either one) as one of the keys, it will cycle both ways, so there is not a need to use both keys for this function. This is a nice one to put there as it is next to the Doppler keys for workflow. CW and Pointer are new with the 2.0 release. Display SIEMENS Healthineers Display Monitor DGC General 1 Screen Saver Delay 20 minutes V DGC Invert with Image Invert System Power Off 2 hours DGC Curve Display Time Out General 2 Brightness 44 DGC Control Current Image Depth Custom Keys Display Patient Banner Patient Data · Age Date of Birth * Exam Doppler / M-Mode Soft Keys Bypass Cursor Mode Doppler Frequency/Velocity Frequency · Velocity Save Cancel HC US MSC S SET Page 13 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Display The System has a screen saver built in. The default is 20 minutes, but can be changed using the drop down menu with the touch pad. The system boots up quickly ( approximately 30 seconds), so to reserve power and battery, set-it to automatically power off Never and 1-3 hours. The default is 2 hours. The monitor brightness defaults to 44, but may be to be slightly changed in bright light situations. .Note: Factory-defined image presets were created using default settings for the brightness of the monitor. Adjusting the brightness may affect the image optimization intended by the factory-defined image presets. For consistency in image reproduction, consistent quality in the image display, and reduced potential for image quality issues, adjustments to the brightness and contrast of the viewing monitor should be made prior to adjusting the print quality of installed documentation devices. The DCG curve display can be set to time out or stay on or off. The DCG control can adjust the DCG controls for the current image depth or maximum depth. The Patient banner can be adjusted to display date of both or age depending on customer preference. The default is age. To bypass the Doppler/M-Mode cursor, check “Bypass Cursor Mode”. The default is “non-checked”. Doppler Frequency or velocity display is selected here. . Exam SIEMENS Healthineers . Exam Exam Abdomen Abdomen General 1 Freeze Behavior OB Early OB Seamless Dual GYN Off General 2 MSK 2D/M & 2D/Doppler Di Breast ID Trace Custom Keys Small Parts Doppler Update Style PV-Arterial / D-L Doppler Auto Stat CV Display PV-Venous D Off Doppler Trace Orienta Renal ve Baseline Below Baseline Both Exam EM Auto Invert of Color V Pelvis Off Soft Keys Auto Invert of Color and Spectrum . On Off M&R 2D Steer with Cursor on · Off Physio Management 3 Lead ECG Display Aux ECG Display Report Aux Signal Display Respiration Display Text Annotation Indicate R-Wave . Lead Type I Lead Type II Save Cancel HC US MSC S SET Page 14 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: The Exam Menu allows the user to set-up many different parameters for workflow. These can be set-up for the specific application, so these are not an all system functionality Freeze Behavior: Once the image is frozen, the behavior of the touchpad and soft key menu will default to the selection. The default is “Cine”, but other choices are text, caliper, arrow or body marker. The most commonly used is Cine. Seamless Dual: For linear transducers, turning this on allows for easy match-up of a thyroid, breast, vascular exam. The default is “Off.” 2D/M & 2D Doppler display allows the choices for ratio of mode to image. Depending on the customer preference, this may be different depending on the application. Doppler Update Style: Doppler Auto Stat: This controls the option to have auto stats always on or off with the choice to activate at the action keys when needed. The default is “Off” Doppler Trace Orientation: Auto Invert of Color with Auto Invert of Color and Spectrum: 2D steer with Cursor: Physio Management: Exam SIEMENS Healthineers Exam Freeze Behavior Display Size Doppler Update Abdominal / Renal Caliper 1/2 2D - 1/2 Trace 2D Frz / D - Lv Emergency Med Arrow 1/3 2D - 2/3 Trace 2D-Lv / D - Lv Vascular Text 2/3 2D - 1/3 trace D-Lv / 2D - 1s Small parts Cine Side by Side D-Lv / 2D - 2s Breast / MSK Body Marker D-Lv / 2D - 3s OB / Early OB D-Lv / 2D - 4s Neo Head D-Lv / 2D - 8s Cardiac / Ped Card HC US MSC S SET Page 15 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: These are the drop down choices for the keys in the Exam menu. Soft Keys SIEMENS Healthineers * Soft Keys 2D Lv 2D Frz CLV | CFIZ | PLV | PFIZ | MLV | MFrZ | DLV | D FIZ General 1 General 2 Physio Settings Custom Keys Display Biopsy Exam Soft Keys << Steer Steer >> MultiHertz M&R U/D Flip Dyn R UR Flip Edge Report Map Tint Text Annotation Full Size Trapezoid Persist CENTER Tx Power Dual Image Presets Adv. SieClear DTCE MultiHertz THI Focus Position Default Clear Page Default Clear Page Clear All Pages Save Cancel HC US MSC S SET Page 16 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: The next menu is programming the Soft Keys. The system allows for maximum flexibility for the user. As noted by the star, these can be programmed for each mode. From 2D, Color, M-Mode, Doppler live to each of these modes frozen. Selecting each one will allow the placement of the key. This is for all applications function. Soft Keys - 2D SIEMENS Healthineers Soft Keys MultiHertz * 2D LV 2D Frz CLV| CFIZ PLV P Frz M LV M Frz D LV D Frz Dyn R Edge Map Physio Settings Tint Persist Tx Power Biopsy << Steer Steer >> THI MultiHertz Focus Position UR Flip UID Flip Dyn R Focus Number Map Tint Persist CENTER DTCE Full Size Trapezoid Tx Power Adv. SieClear Dual Image Presets Adv. SieClear DTCE MultiHertz THI Focus Position Image Presets Blank Default Clear Page Default Clear Page Clear All Pages Save Cancel HC US MSC S SET Page 17 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: This is an example of the choices for 2D image parameters. Each key has the same choices. Soft Keys - 2D Left Panel Control Keys SIEMENS Healthineers Soft Keys 2D LV 2D Frz | C | CFr | PLV | PFZ| ML|MFr| | D FIZ Full Size Dual Physio Settings U/D Flip UR Flip Biopsy Trapezoid Biopsy Clip Setting: Steer > < < Steer Biopsy Steer >> Physio Settil Blank UID Flip Full Size Trapezoid CENTER Clip Settings Dual Image Presets |Adv. SieClear DTCE MultiHertz TH Focus Position << Steer Steer >> Physio Settil Default Clear Page Default Clear Page Clear All Pages Blank Save Cancel HC US MSC S SET Page 18 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: This is an example of the choices for 2D image parameters. Each key has the same choices. Measurements and Reports SIEMENS Healthineers M&R General 1 -Measurement and Report Configuration General 2 Cardiac Custom Keys Display Measurement Method >> Exam Measurement Order >> Soft Keys Display Items >> M&R User-Defined Measurement Labels >> Report Comments Library >> Text Annotation General Configuration >>> Body Marker Anatomy Assessments >> Wireless HC US MSC S SET Page 19 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Measurement Order: The Next menu choice is for Measurements and reports. A separate presentation will be given on this option. This is also programmable by the application and is more extensive in applications with dedicated measurement packages. Using the Touch pad, scroll to the >> and see the choices for each menu item. Measurement Method SIEMENS Healthineers M&R >> Measurement Method : Abdomen * - Default Measurement Method by Mode 2D Mode Distance Measurement Method >> M-Mode HR Doppler Velocity Measurement Order >> -Default Measurement Method User-Defined Measurement Labels >> Area Ellipse Comments Library >> Volume 3 Dist % Stenosi Area - %Stenosis General Configuration >> 2D Mode Ratio Distance Ratio M-Mode Ratio Distance Ratio Anatomy Assessments >> Flow Volume Area - Flow Volume PI Trace PI Auto Default Previous Save Cancel HC US MSC S SET Page 20 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Measurement Method: The first choice is measurement Method. Each section has a drop down menu. These are the default values for this application. Measurement Order SIEMENS Healthineers . M&R >> Measurement Order : Abdomen Mode 2D-Mode 2D-Mode M-Mode Measurement Method >> Measur Measurement Order Doppler TEST * Measurement Order >> User-Defined Measurement Labels >> Add Remove Comments Library >> Move U Move Down General Configuration >> Anatomy Assessments >> Default Previous Save Cancel HC US MSC S SET Page 21 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Measurement Order: Depending if the application has an associated measurement package with it, this could have some measurements already populated. In this case, there is no measurement package, so Test was added as a user defined. Display Items SIEMENS Healthineers M&R >> > Measurement Results : Cardiac General Default Measurement Title 2D Mode None M-Mode None Measurement Method >> Doppler None Measurement Order >> Measured Results Direct Average Display Items >> HR Measurement Method · ECG Doppler / M-Mode O Auto Stat Display of Measurement Labels Measurement Type · Last Used User-Defined Measurement Labels >> Display AOLA AOd/LAS Y LAS/AOd Comments Library >> Default RAP 10mmHg General Configuration >> eSie Measure Settings Anatomy Assessments >> eSie Measure On · Off Heart Cycle to Display M-Mode 3 Doppler W Default Previous HC US MSC S SET Page 22 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 User-defined Measurement Labels SIEMENS Healthineers M&R >> User-Defined Label : MSK 2D Mode User-Defined Measurement Label BURSA Measurement Method >> User-Defined 1 Label Name Measurement Method Distance Distance Measurement Order >> Delete Current 2D Mode Label User-Defined Measurement Labels >> M-Mode User-Defined Measurement Label User-Defined 1 Label Name Comments Library >> Measurement Method Distance General Configuration >> Delete Current M-Mode Label Anatomy Assessments >> Doppler Mode User-Defined Measurement Label * User-Defined 1 Label Name Measurement Method Velocity Delete Current Doppler Label Previous Save Cancel IC US MSC S SET Page 23 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: The next menu choice is creating user-defined labels for the different measurements modes. The number of user-defined available to create is : 20 for 2D , M-Mode and Doppler. The M-mode and Doppler was increased with this software. The type of measurement is available as well depending on the mode. General Configuration SIEMENS Healthineers M&R >> General Configuration : Abdomen - Caliper Configuration Caliper Default Position Center Measurement Method >> Caliper Size Large Measurement Order >> Caliper Pattern Measurement Results Background Con Off User-Defined Measurement Labels >> -Measurement Result Display Comments Library >> 2D Mode Bottom Doppler / M-Mode Bottom * Top Bottom General Configuration >> Left Right Anatomy Assessments >> Default Previous Save Cancel HC US MSC S SET Page 24 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: This is a common page where the Caliper placement, size and type can be selected.. These are applications specific. One of the key things that is accessed in this menu is where the measurements are displayed. Many users have preferences regarding this. Anatomy Assessment SIEMENS Healthineers M&R >> > Anatomy Assessments : Abdomen Item Selections Measurement Method >> Items List *Liver echogenic Liver Measurement Order >> User-Defined Measurement Labels >> Comments Library >> Add Remove General Configuration >> Anatomy Assessments >> Default Previous Save Cancel HC US MSC S SET Page 25 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Some general comments regarding anatomy clinical findings can be added here. This is especially useful for General Imaging for Abdominal, Thyroid and Breast. Report SIEMENS Healthineers Configuration Report Report Order Print Preview Qview General 1 General 2 General 1 Custom Keys Site Information Display Exam Logo Text 1 General 2 Soft Keys M&R Font Custom Keys Report Text 2 Text Annotation Font Body Marker Display Wireless Load Remove Text 3 Network Font Exam Service Clip Store Archive Soft Keys Authorization -Signature Printer M&R Barcode Utility Add Remove Report Text Annotation Body Marker Wireless Save Cancel HC US MSC S SET Page 26 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Report: In the report section, this is where the customer logo is added. One it is on the USB device and attached to the system, pressing the Load button will find the folder and allow it to be uploaded. It must be jpeg. And not in a folder. Physician signature can be added and removed here. Additional information and exercises will be in the Measurement and Reports presentation and workshop. Text Annotation SIEMENS Healthineers Text Annotation Exam Small Parts General 1 Annotation Library TextA Text B * General 2 Enable Auto Complete Custom Keys Available Annotations Selected Annotations Display Left Thyroid Right Isthmus Sagittal Lobe Exam Trans Medial Coronal Lateral Superior Soft Keys Modify Mid Inferior Add Anterior M&R Posterior Remove Node Solid Report Cystic Calcification Text Annotation Anechoic Echogenic Hypoechoic Body Marker ICA ECA CCA Wireless Jugular Parathyroid Network Default Save Cancel HC US MSC S SET Page 27 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: The user can Customize annotation libraries for selected exam types and user-defined exams. They can enter user-defined annotation or modify selected annotation. From the menu, the user can Add/Delete selected annotation and Move Up/Move Down the selected annotation to order based on their protocol. Body Marker SIEMENS Healthineers . Body Marker Exam Small Parts General 1 Available Body Markers Selected Body Markers General 2 Custom Keys Display Add Exam Remove Soft Keys M&R Report Text Annotation Body Marker Wireless Default Network Save Cancel HC US MSC S SET Page 28 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Body Markers: The user can Customize body marker libraries for selected exam types or user-defined exams The menu displays a library of pictograms for selected exam type or user-defined exams. All body markers are in the library, so any can be added to the specific exam. The User can Add/Delete selected pictogram from displayed library and Move Up/Move Down the selected pictogram within the library for order of placement depending on their protocol. Wireless SIEMENS Healthineers . Wireless Config Wireless Network General 1 General 2 Wireless On/Off @ On off Custom Keys IP Address Settings Available Networks Display O DHCP Static IP Address SSID Connection | Signal | Security | Saved Exam IP Address: 0 .0 .0 .0 Soft Keys Subnet Mask: 0 .0 .0 M&R Default Gateway: 0 . 0 . 0 . 0 MAC Address: 3C - A9 - F4 - AB - 93 - 44 Report Clear Apply Add Text Annotation Refresh Edit Connec Body Marker Status: Searching for wireless networks ... Wireless Network Service Clip Store Save Cancel HC US MSC S SET Page 29 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Wireless sends data, such as studies, images, and clips to a network location over a wireless network ƒ Requires approved wireless adapter, purchased separately ƒ Antenna 2.4GHz, 5.8GHz ƒ Wireless is a standard feature. Network SIEMENS Healthineers Network General 1 Local Host Storage Server | Worklist Server | Network Export Server | General 2 Default Show Network Status Custom Keys Display Exam New Edit Delete Soft Keys O DHCP Static IP Address M&R Alias: Default IP Address: 0 . 0 . 0 . 0 Report AE Title: Default Subnet Mask 0 . 0 . 0 . 0 Text Annotation Host Name: Default Default Gateway: 0 . 0 . 0 .0 Body Marker MAC Address: 00-13-95-15-FA-CC Wireless Media Type Hardware Default Duplex Mode Hardware Default Network Service Clip Store Archive Authorization Save Cancel HC US MSC S SET Page 30 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Network Setting up the DICOM network may be done by the local service team. Should it be required to complete, review the manual for correct set-up. It will be important to work with the local IT department onsite to get the information neededto interface. DICOM is an option on the system. Note: DICOM printer is not supported. Network Export is an excellent way to send reports over to a PC when a DICOM network is not in place. Service SIEMENS Healthineers . Service General 1 -System Information General 2 ACUSON P500 KT PP3018_V 0.9.4 Custom Keys SIN PP3018 Display Version: VADOA (0.9.4_p1) Exam System: VA00A (0.9.4_p1) Soft Keys System Internal: 0.9.4_p1 S/N: PP3018 [20141128 12:43:01] M&R P/N: 10785577 [20141128 12:43:01] OS Configuration: 0.0.6 Report Framework: 0.10.5  Platform Software: 0.18.8  Text Annotation PIMS: 0.8.35  UBE: 0.8.20  Body Marker Wireless Network Service Service Clip Store Remote Archive Remote Application Support Authorization Printer Restore Factory Settings Barcode Utility Save Cancel HC US MSC S SET Page 31 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Service: This is where the user can access the service menus. It is also an excellent place to identify the system serial number and software version. Service SIEMENS Healthineers 2D System Serial Number : PP4294 7/10/2014 10: 14:50 2D System Serial Number : PP.4294 07/10/2014 10:15:40 CH5-2 OB OB Tx Power 100% Tx Power 100% MI 0.67 MI 0.67 TIS 0.4 TIS 0.4 TIB 0.4 TIB 0.4 ASC 1 ASC 1 DTCE Med SIEMENS DTCE Med Dyn R 75 dB Dyn R 75 dB Biopsy THI 3.6 MHZ 0 dB Biopsy THI 3.6 MHZ 0 dB 23 fps 23 fps netnewer LAN setup - proxy server << Steer|Stoer>> Please specify address and port no. of your proxy server << Steet Steff >> O Address: netviewer Port: IF Save settings The session was dosed. UR Flip WD Flip abort UR Flip UND Flip Close U Powered by www.netviewer.com Full Size Trapezoid Full Size Trapezola 17 cm Dual 17 cm Dual Delete THT Adv . Edge Focus Nu ... SieClear DICE Dyn R MultiHertz Position Te Howe TEQ Gain Image Pre- R/S Focus TEQ Gain Persist FOCUS Poi, Med 75 dB 3.6 MHZ Map THI SieClea DICE Dyn R MultiHertz How Edge Focus Nu .... FOCUS PorL. Persist Focus Posi ... Med Position Position 75 dR. 3.6 MHZ Map Focus NU ... Focus Posi ... HC US MSC S SET Page 32 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Service: Remote and Remote Applications Support This additional Choice for SRS ( Siemens Remote services) are available for set-up here. SRS us standard with the first year of warranty, but is part of a service contract for long term use. Please Note: SRS may not be available in all countries. Check with your local country to ensure availability before discussing. Specific information will be needed. The user will have to be connected to the internet for this to work and specific time availability may vary. If the user wants to do remote service, it is important to work closely with your local service engineer to ensure this is set-up properly. SRS is the enabling connection for eSie Link remote access. Clip Store SIEMENS Healthineers Clip Store General 1 General 2 Compression: .Low Medium High Custom Keys Display Capture Type: · Prospective Retrospective * Exam Soft Keys Clip Length: Time 3 sec M&R Beat Report beats Text Annotation Body Marker R-wave only R-wave delay: 0 msec Wireless Network Service Clip Store Archive Authorization Printer Barcode Utility Save Cancel HC US MSC S SET Page 33 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Clip Store: This allows the user to : ƒ Choose the compression for best for image quality and storage size ƒ Set for a time capture from 1-60 seconds ƒ The capture can be prospective or retrospective (retrospective capture adds in a few seconds before press of clip store, Prospective starts at press of Clip Capture.) ƒ R-Wave Only ECG Trigger Type For procedural exams, retrospective capture can be useful to capture the needle. Just set the time frame slightly longer. This is a set for all exams. Archive SIEMENS Healthineers . Archive General General 1 Options for Deleting Archived Studies General 2 Delete Study History after Study is Deleted Custom Keys Display Manually Delete Study History after Study is Deleted Exam Soft Keys Keep Study History after Study is Deleted M&R Report Show Deleted Studies in Browser Text Annotation Body Marker Wireless Network Service Clip Store Archive Authorization Printer Barcode Utility Save Cancel HC US MSC S SET Page 34 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Archive: This allows the user to determine how studies will be archived and if deleted studies are shown on the browser. The default is “Delete Study History after Study is Deleted”. Authorization SIEMENS Healthineers ' Authorization General 1 Change Password Change .. Administrator Login ... General 2 Custom Keys Accounts Administrator Disabled Display Exam Reset ... Delete New Soft Keys Source Media [G:] AMY M&R Import Export Report Text Annotation Authorization Policy Body Marker Login Required Login Screen Message Change ... Wireless Retry Attempts and Period 3 Attempts 11 Seconds Network Auto Logout (Screen Saver) Service Expiration days Clip Store Archive Password Policy 8 Minimum Characters 2 Numeric Characters Authorization 2 Special Characters 2 Uppercase Characters 2 Lowercase Characters Printer Barcode Utility Save Cancel HC US MSC S SET Page 35 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Authorization When accessing the Authorization screen for the first time, the system prompts a set up of an administrator account. Passwords access is set-up here. Printer SIEMENS Healthineers . Printer General 1 Printer General 2 PC Printer Sony UP-D25MD Status Custom Keys Display USB B&W Printer Status Exam Status Soft Keys USB Color Printer M&R Install ... Report Text Annotation Thumbnails Printer Body Marker . None Wireless Sony UP-D898MD/X898MD Network PC Printer Sony UP-D25MD * Service MITSUBISHI P95D Clip Store USB B&W Printer MITSUBISHI CP30D DW(USB) Archive USB Color Printer Authorization Printer Barcode Utility Save Cancel HC US MSC S SET Page 36 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Printer: This is the location to select the printer to be used as well as guide through the installation procedure. This is usually done at the time of the service installation and performed by service. Each user should practice adjusting the printer setting and loading paper. The drop downs menus have the printers listed in the box next to the star. Select the desired choice to install. Barcode SIEMENS Healthineers Barcode Settings Connection Status Disconnected General 1 General 2 Sample Text Offset Length Custom Keys Scanned Data Scan a barcode Display Use Separator Exam Soft Keys Offset Length Order M&R Last Name Set 65 Report First Name Se 65 Text Annotation Patient ID Set 65 Body Marker Wireless Performing MD Set 65 Network Service V Sonographer Set 65 Clip Store Accession No. Set 65 Archive Authorization Printer Barcode Utility Save Cancel HC US MSC S SET Page 37 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Barcode: Setting up the barcode is done in this screen. The user can select the information that needs to be scanned here. The default is “not checked”. Step by step directions are available in the manual. The barcode device is connected to one of the USB ports. This is the same barcode device that is used on the ACUSON X600, ACUSON X700 and ACUSON X300 PE systems. Utility SIEMENS Healthineers Utility General 1 ...... Clip Playback General 2 . Play Only Active Clips Custom Keys Play All Clips Loop Aligned Display Exam Soft Keys Slide Show M&R Image Duration (sec) * 3 Number of Replay Loops Report Text Annotation Body Marker Wireless Network Service Clip Store Archive Authorization Printer Barcode Utility Save Cancel HC US MSC S SET Page 38 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Utility This is where the duration of the Slide show is set. The actual slide show feature is set-up in the Review Screen, but the duration is set here. In addition, in review, set-up whether all clips or just a selected clip plays back. The default is “ Play only active clips”. Stress Echo SIEMENS Healthineers . Stress Echo Protocols Configuration Pharmacology Rest General 1 Pharmacology Continuous LOW General 2 Bicycle Peak VV V V Custom Keys Bicycle continuous Post Display V Exam Treadmill Soft Keys Treadmill Continuous M&R Report Text Annotation Copy Add Rename Delete Body Marker Wireless Protocol Settings -Review Settings Network Service Y ROI On Display Format : 4x4 Clip Store Archive Allow Selection of Multiple Clips per View Default Playback Mode : Full R-R Authorization Printer Default Playback Speed : 100% Barcode Utility Default Clip Sorting : . By View O By Stage Stress Echo eSieScan Export Import Default Save Cancel HC US MSC S SET Page 39 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: Stress Echo This is where the user sets-up the protocols for Stress echo and the review settings. eSie Scan™ Workflow Protocols SIEMENS Healthineers eSieScan Protocols Configuration E Routine Echo Protocol Name PLAX PLAX General 1 PLAX General 2 PLAX Rename Custom Keys PLAX Display RVIT RVIT Automatically ends when all views completed. Exam RVIT -At End Protocol- Soft Keys PSAX AV M&R PSAX AV Remain in acquisition Report PSAX IAS PSAX TV Enter Review Text Annotation PSAX TV Body Marker RVOT Enter Report Wireless RVOT RVOT PV Network PSAX LV BASE Service PSAX LV MID PAP Clip Store PSAX APEX Apical 4ch Archive Apical 4ch Add Protocol Authorization Apical 4ch Printer Apical 4ch Add View Apical 4ch Barcode Apical 4ch Copy Utility Apical 5ch Default Stress Echo Apical 5ch V Delete Apical 2ch eSieScan Export Import Save Cancel HC US MSC S SET Page 40 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Speaker Notes: eSie Scan Workflow protocols for Cardiology are customized here. Protocols are not available for other exam types. Summary SIEMENS Healthineers . Access the Configuration Menu Identifying the Configuration Menu Choices Making Changes to the Configuration Menu Review Configuration Menu HC US MSC S SET Page 41 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Questions HC US MSC S SET Page 42 | Unrestricted C Siemens Medical Solutions USA, Inc. 2017 Questions?
- ACUSON P500