ARTIS icono biplane VE20 Store Direct Positions Video
This video will demonstrate how to store direct positions on the ARTIS icono biplane VE20, on the Pilot Control Module in the exam room.
Welcome to the Artis Icono biplane store direct positions video. In this video you will learn how to store frequently used system positions as direct positions on the pilot control module PCM. There are three keys surrounding the left joystick on the pilot control module for storing direct positions. Direct position 1, two and three. Here you can store your frequently used system positions so that you can access them easily during the examination. To store a direct position, move the See Arm system to the desired position. Direct positions are intended to provide an identical view and image impression only. The viewing direction, zoom and filter settings are restored. Press the store key at the front of the left joystick. The LEDs on the direct position keys flash. Select the key on which you would like to store the direct position. When the position is stored, a message appears on the control room display and on the exam room display. Additionally, an audible sound will be heard. The position will remain stored until overwritten. To move the system to the direct position, press the direct position Key, 1, two, or three. Press down on the joystick and deflect the joystick away from you. Please note it is also possible to move the system to the direct position by pressing down and holding the direct position key.
@ Siemens Healthcare GmbH, 2019 Please note that the learning material is for training purposes only! ARTIS icono biplane For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be SIEMENS used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. Store direct positions Healthineers The Operator Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. The information in this material contains general technical descriptions of specifications and options as well The reproduction, transmission or distribution of this training or its contents is not permitted without as standard and optional features that do not always have to be present in individual cases. express written authority. Offenders will be liable for damages. Certain products, product related claims or functionalities described in the material (hereinafter collectively All names and data of patients, parameters and configuration dependent designations are fictional "Functionality") may not (yet) be commercially available in your country. Due to regulatory requirements, and examples only. 0 01 the future availability of said Functionalities in any specific country is not guaranteed. Please contact your о 01 local Siemens Healthineers sales representative for the most current information. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. i11t
- store position