ARTIS icono floor VE20 3D Acquisition Video
This video will demonstrate the workflow for 3D acquisitions on the ARTIS icono floor VE20.
Welcome to the Artis Icono floor 3D acquisition video. In this video you will learn the workflow to perform 3D acquisitions. First, select the 3D acquisition protocol. If your current exam room display is configured to display the 3D wizard, you can use the 3D wizard to select the protocol and for acquisition guidance, you select the 3D wizard from the onscreen menu. Press the key on the pilot control module to open the onscreen menu. Deflect the jogwheel to select the general 3D sharing menu, an select the 3D wizard. Alternatively, you can open the 3D wizard from the touch screen in the examination menu. Select the 3D acquisition protocol. Please note if your current procedure contains a 3D protocol, it is also possible to select the 3D protocol manually from the examination menu on the touch screen. 3D acquisitions are performed with the system position that the table head side. To move the system to the starting position, press down on the left joystick and deflect the joystick away from you. The system moves to the start position for the 3D acquisition. When positioning the system, follow the guidance on the exam room. Display guidance messages appear for all steps of the acquisition workflow. It is a requirement for 3D acquisitions that the region of interest is positioned in the isocenter position. The region of interest in the center in the AP position perform fluoroscopy briefly. Using care position, adjust the table lanja Tude and Leanne transversely so that the region of interest is in the center of the field of view. Depending on the region of interest, collimation may be applied, but only in the top to bottom direction. Slab acquisition using side to side collimation must not be performed as the image quality is impaired by truncations. To position the region of interest in the lateral position, press on the stand. See arm joystick and deflect the joystick away from you. Perform fluoroscopy briefly using care position. Adjust the tabletop height so that the region of interest is in the isocenter. The Optional Laser Cross might be of help during this step. When you are satisfied that the region of interest is positioned within the isocenter, you can continue with the 3D workflow and enter the test phase two prepare, move the system to the start position for the test phase. Press on the stand. See arm joystick and deflect the joystick away from you when the system is in the start position for the test phase, a message appears on the exam room display. It is important in the test phase to ensure that the area of rotation is free from possible collisions. Move any equipment away from the rotation zone, including the exam room display to start the test phase, press down and deflect the left joystick away from you. The See Arm will rotate around the region of interest. This test is performed at a slower speed than the later acquisition to allow a better reaction to possible required changes. Keep the joystick deflected. The test phase will end automatically. For automatic setting of acquisition parameters, you are prompted by the system at the end of the test phase to perform fluoroscopy when the test phase is complete, the system is ready for acquisition before you start the 3D acquisition. Ensure that you are prepared for the injection of contrast media if required. See arm rotation depends on the selected protocol. For example, in Der acquisitions the system will perform one. See arm rotation. If an injector is armed, the contrast agent will be injected. DS acquisitions will involve three rotations of the C arm. The mask run the return rotation no X Ray and the fill run the injector will be started for the fill run. The system will wait the time that is set in the X Ray delay. Some acquisition protocols involve a bolus watching phase. After the mask run. For example, Dina Pve Nuro acquisition can be released by the Hand Switch or the foot switch. The contrast medium injection can be synced with the acquisition, or you can manually trigger the injection of contrast media using the injector interface during the 3D acquisition. As the C arm rotates around the region of interest, the 2D images are shown on the exam room. In control room displays, this allows you to see the anatomical structures, an monitor. The injection of contrast media. Continue to release X Ray hand Switch or footswitch until you are notified by the system that the 3D run is completed. An audible signal indicates that the acquisition is complete. When the 3D acquisition is finalized, the 3D acquisition program is automatically deselected, and a 2D acquisition program is selected by default. If configured, the system can be moved back to the working position prior to entering the 3D acquisition workflow.
Head & Neck Patient. Siemens Healthineers High Contrast Vessel Tree 4677843345 Patient. Siemens Healthineers 4677843345 Information 3D Wizard Wizard Save dookmark Save Angulation Bookmark DSA acquisition Reset All Orientations @ Siemens Healthcare GmbH, 2019 Review Review Dyna4D Hide All Objects Save Snapshot Please note that the learning material is for training purposes only! 3D Reconstruction... ARTIS icono floor For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be SIEMENS used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. 3D Acquisition Healthineers ' The Operator Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. PATIENT. Siemens 1.0 1.0 Healthineers Patient. Siemens Healthineers Ref Image +1- min 0.0 KV Gy 2% 20/0 2% 3% 1465.2 3D: Deflect joystick to move system to ISO A:3D test phase: Start fluoro to complete C-Arm Position 3:41 PM 0.0 KV 0.0 mA Note: Some functions shown in this material are optional and might not be part of your system. The Examination ARTIS AP position DSA 4 CNR (4.0 f/s) FL Neuro (10.0 PIs) 5.36 pGym2 8.80 pGym2 1465.2 Navigate Image +1- test phase lay ack 50.00 mGy ooo.l min 0.0 mA information in this material contains general technical descriptions of specifications and options as well 0.00 mGy O.OO mGy 50.00 mGy 000.3 min 000.3 min ooo.l min Pause Image +1- Scene +1- Series Descr. as standard and optional features that do not always have to be present in individual cases. Examination Display Measure Field C-Arm Position AP position Table Certain products, product related claims or functionalities described in the material (hereinafter collectively Scene Length "Functionality") may not (yet) be commercially available in your country. Due to regulatory requirements, Detector RDMP & Overlay 3D Acquisition DSA acquisition the future availability of said Functionalities in any specific country is not guaranteed. Please contact your ARTIS Block x.Ray Case Flows Tools local Siemens Healthineers sales representative for the most current information. Examination Display 3D Wizard Wizard Favorites ARTIS Table Sys. Pos, Pause Favorites
- 3D acquisition