ARTIS icono VE20 Load Images 3D Sharing Control Room Video
This video will demonstrate how to load images to 3D Sharing on the ARTIS icono VE20 in the control room.
Welcome to the Artis Icono load images to 3D sharing and to the viewer video. In this video you will learn how to load 3D images to 3D sharing so that the images can be shared on the exam room display. Additionally you will learn how to load images to the viewer during the live examination. You use 3D sharing to share the same view of the 3D volume in the control room. And in the exam room. Check that 3D sharing is selected on the control room display following a 3D acquisition. The images are automatically loaded to 3D sharing to loaded. Previously acquired 3D volume from the current examination. Select the image directory. Take care that you open the image directory in 3D sharing and not in the live case segment. Double click on the required images in 3D sharing only 3D volumes can be selected. The images are automatically updated on the exam room display. The images on both displays are identical and synchronized with each other. That means that if you update the image on one display, the image is automatically updated on the other display. To load images from a previous examination for the current patient, for example, a previously acquired CT or MRI to 3D sharing. Open the patient browser. Take care that you open the patient browser in 3D sharing and not in the live case segment. The volume now appears in the 3D sharing segment on the control room display and on the exam room display. To View images on the control room display only, you open the images in the viewer. To change the segment to the viewer, click on the dropdown menu and select the viewer. To open a series in the viewer, open the patient browser when opening the browser, take care that you open the browser in the viewer and not in the live case segment. Double click on the desired series in the viewer, both 2D and 3D series can be selected.
1 program (15 f/s) I.CARD program (15 f/s) CD CT 3D PATIENT, Siemens p Gym* 49 cm 10 cm 235 cm program (VFR) 6.DSA program (VFR) (40). 10 I.CARD Exam set (40). 10 I-CARD Exam set (40). 10 1234567. I.CARD Exam set (40). 10 1234567. I,CARD Exam set mGy 'JGym7 pGym7 49 cm 235 cm 49 cm 10 cm 235 cm 5 S_nuoro normal (15 pjs) 5_nuoro normal (15 pjs) 0 5_nuoro normal (15 pjs) 5_nuoro normal (15 pjs) Patient Browser 36 3D 3D PatientASiemens Healt... Viewer' anng armg 30 test2 (1) mage Viewing Review loaded or acquired volumes. ed or acquired volumes. Live A New acquisitions Will replace existing volumes. New acquisitions replace existing volumes. lions Will replace existing volumes. Patient CT Refl A 3D Patient, Siemens Healthineers D Patient, Siemens Healthineers D Patient, Siemens Healthineers • 3D Patient, Siemens Healthineers 0020329995 020329995 0020329995 020329995 3D Series Results Series Description Modal ny Date Archived Study Description Radiation Dose 3D Patient, Siemens Healthineers PM 1.3.1221107.5.... Refl A w•2032gggs 12/11/2015 3D Patient, Siemens Ref2 B •11/11/1978. M 1234567 3D Sharing PM) AM 12118/2013 PM 12±52/ PM • 3D Patient, Siemens Healthineers 10 cm 49 cm cm Coro Clear minus Viewer2 A 12/1112015 AM PM 3D PATIENT, Siemens 3D Patient, Siemens Healthineers Viewerl B 12/11/2015 18 room 0020329995 Healthineers ' CIA 22252 PM < Review* C Review> < Review* < Review> < Reviews 12/11/2015 AM SL SL 54 364 Clear minus 184.108.40.2067.5_ MPR MPR 112 LAO go / o 21 20 Coro Clear minus CT Instances 10 cm 49 cm Scene CARD 7.5 CARD Scene CARD 7.5 f/s CARD 7.5 CARD C Review> < Reviews < Review* CReview> SL 548 30 364 36 MPR 118 112 29 294 O CAUD go
- 3D Sharing