Artis Systems VD12: Image Review Touchscreen Control Video
This video explains how to review 2D images using the buttons on the touchscreen control (TSC) and the mouse joystick.
Welcome to the Artis 2D image review on the touch screen control cvideo. In this video you will learn how to review 2D images using the buttons on your touchscreen control and the mouse joystick. You find the buttons for 2D image review on the borders of the task card stack on the touch screen. Please note that the buttons available on the touch screen and their position depends on your system configuration. Press the button to open the image player. Press the plus or minus buttons to scroll through the scenes. If required, press the button to open the scene directory. Use the mouse joystick to navigate through the directory. Press on the left mouse button to select a scene for review. The scene or image displays in full screen. To set the review frame rate, deflect the mouse joystick. The more that the joystick is deflected, the faster the scene is played back. Release the mouse joystick when the desired speed is reached. To stop a loop from playing, deflect the mouse joystick briefly. Deflect the joystick to the left to scroll through the scene backward frame by frame. Deflect the joystick to the right to scroll through the scene forwards frame by frame. Alternatively, you can use the buttons in the image player to scroll through the scene frame by frame. If configured in the layout, press the loop through all scenes button on the touch screen to review one scene after the other. To review reference images, press the button to open the image player for reference images. You will find the buttons for post processing images in the image task card. The buttons are grouped by function. The functions available depend on your touchscreen configuration.
The reproduction, transmission or distribution of this training or its contents is not permitted without @ Siemens Healthcare GmbH, 2020 express written authority. Offenders will be liable for damages. Please note that the learning material is for training purposes only! All names and data of patients, parameters and configuration dependent designations are fictional and For the proper use of the software or hardware, please always use the Operator Manual or Instructions for examples only. na 'Exam Image'] Quant' // Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be used ngs Exam Irnage Quant 4 40 as training material only and shall by no means substitute the Operator Manual. Any material used in this All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Artis Systems training will not be updated on a regular basis and does not necessarily reflect the latest version of the 3D Head < Default> SIEMENS software and hardware available at the time of the training. FL(-) Neuro DR Overview Adjust The Operator Manual shall be used as your main reference, in particular for relevant safety information like smooth warnings and cautions. Healthineers 4 p/s Image Review Touchscreen Control REF Applications DSA Note: Some functions shown in this material are optional and might not be part of your system. REF Certain products, product related claims or functionalities described in the material (hereinafter collectively Tools ROMP ROMP ROM p "Functionality") may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information.
- artis q.zen
- artis qzen
- atris qzen