Artis Systems VD12: Post-processing Control Room Video
This video explains how to review and post-process images from the Postprocessing task card on the control room user interface.
- 0:28: Image Review
- 2:42 Windowing
- 4:00 Edge Enhancement
- 4:38 Image Shutter
Artis 2D post processing video. In this video. you will learn how to review and post process images from the Post Processing Task card on the control Room user interface. In the Post Processing Task card you will find icons for image review at the top of the task card. You will find further icons for image review and post processing in the view image, and adjust subtests cards. You can also access functions from the view and image menus in the main menu. Additionally, you can find some functions in the context menu. You access the context menu by right clicking on the image. You can select the type of image for review, select the arrow to select an option from the dropdown list. Use the plus or minus buttons to scroll through scenes or images. Alternatively, you can press on the scene plus and scene minus keys on your keyboard. You can open the scene directory, click on the icon to open the directory. The system switches to 4 by 4 display mode. If more than 16 scenes or images are stored, use the scrollbar if required. For scene review, you can use the review control. The acquisition mode and the frame rate are displayed in the top left corner. You can review scenes at the same frame rate as they were acquired, or you can slow down or speed up the playback speed. Click and drag the slider to adjust the frame rate speed. To stop the loop playing, click the Stop preview button. Alternatively, you can stop the loop from playing by clicking in the image area with the left mouse button. If required, click on play to start the loop. To loop through all scenes, select the View Subtests card and select the loop all scenes icon. Alternatively, you can select view loop all scenes in the main menu. With windowing you can adapt the image to display the grayscale region of interest with optimal contrast and brightness conditions. There are several methods available for windowing images in the control room. To window the image using the mouse, press and hold the middle mouse button in the required image. To adjust the window center image brightness, move the mouse up and down. To adjust the window width image contrast, move the mouse left and right. You can also use your keyboard to adjust the window values. Alternatively, you can adjust the contrast and brightness in the adjust subtests card. Select the auto windowing function to ally optimized window values. You can enter the desired window in the field or use the up or down arrows to adjust the value. To reset the window values to the original values, select the home icon. In the adjust sub task card you can also adjust the edge enhancement. Adjust the value as required using the up and down buttons to reset the values to the original value. Select the home icon. In the view test card you will find the icons to change the display from full frame view to scene overview and the icons to shutter the image, invert the image, zoom by a factor of two and pan the image. Zoom to acquisition size and loop through all scenes. To shutter an image, select the shutter icon in the view sub task card. Alternatively, you can select view and shutter in the main menu or right click and select shutter in the context menu. After selection of shutter, the mouse cursor changes. Click and drag the mouse on the edge of the shutter to adjust the shutter edge. To pan the image within the shutter press and hold the shift key on the keyboard and click and drag the image.
Patent mage Qptons Siernens Health. Eat] ent Image Options emens kea Ithineers atiÅnf,eese2 emens Healthineers ese,2 iemens Healthineers Patient, emens Healthineeks+"entGSé2 emens Healthineers ÅÅåent:ease2 emens Healthineéks ÅÅåent:ease2 emens Healthineers+aålent, Siernens Health. Siernens Heath. Siemens Health Siemens Health_ Health. Image emens HealthineéF4PåWénf,VCase2 emens kea Ithineers emens Healthineers+atient, Case2 emens Healthineers +afient, ese2 emens Heålthiheer4pakVentiZåse2 emens HealthineekS lemens Healthineers+åtient, emens Healthlnee emens Healthineers Patient, Siemens Healthineers Patient, Case2 Siemens Healthineers Patient, emens Healthineer4pakVent,Zåse2 remens Healthineers Patient, Case2 iemens Healthineers Patient. Case2 emens Healthinee emens Healthineers emens Healthineers ese2 emens Healthineers+"nt,2 emens Healthinee a e ase2 case2 Case 2 S Heath Heath Siemnens Health. Siemens Health Siernens Heath. emens Case2 Case 2 Case2 iemens Healthineers Patient, Case2 Case2 Zoom 230507 Siernens Health. e ens Healthinee 1234567 34567 Healthine case2 ase2 Heath 234567 Health Case 2 He 234567 1/1/1900 11/1900 *1/1/1900 1/1/1 goo 234567 4567 230507 234567 Shutter * 1/1/1900 case2 Case 2 11 goo 11/1900 1/1/1900 12/11/2015 2/11/2015 11/2015 Shutter Scene 16:15 AM .16:15 AM 16:15AM :15AM 8888 Scene Timer CARD 7.5 RAO 70 CARO 7 CARO 11/2015 58/32 CARD 7St/s CARO 7St1s CARD7St1s CARD 7St1s CARO 7 5 4 - 19/32 :15AM 20/32 10/32 22132 7132 28/32 29/32 27132 19/32 - 20/32 16:15AM 6:15AM - 29/32 CARO CARD 7.5 CARO 7 CARO 27132 22132 89/32 16/32 19/32 2/11/2015 62 sec 1129 sec 196 sec •16 sec 46 sec- 2.46 sec .46 sec .13 sec 191 sec .78 sec .49 sec .00 sec .62 sec 49 sec 13 sec 78 sec 52 sec 46 sec 6 sec 2 sec Mode 2.33 sec CARD 3 sec 6 sec 2 sec 4 sec 62 sec Flip Horizontally - 20/32 CARD 7.5 CARD 7St/s 20/32 STOP Display View OSA DSA View OSA DSA .46 sec 2.33 sec View OSA Invert Shift b. Tools ools Healthineers Healthin 2050 zoom 2300 2252 auto Shutter auto auto Image Scene Timer Scene Windowing opac 116B x 11 116B x MI 116B x Ml 2300 auto Ju 116B x 11 1168% 11 EE 10 1168 HI 1168111 1168 WII 1168 Ml 1168x1f68 1168x1f 1168% If 1168x11 1168k1168 1168M 1 1168x1168 1168% 11 1168 x 11 Coro DDO EE 10 60 RAO 76 DDO 60 EE 11% cm 26 RAO 70 60 23050 230507 wc 20 cm 20 DDO auto cm 26 D 99 w 96 kV 83 W 83 kv 83 W 96 kv 83 k 83 96 w 83 83 mA 820 mA 413 mA413 413 kV 83 W 83 D 99 mÅ413 99 wc 2050 wc 20 mA 413 mA 413 W 83 Coro RAO 76 RAO T RAO 70 2300 23050 RAO O• 2418 cm 26 RAO T RAO 76 DDO 60
- artis q.zen
- artis qzen
- atris qzen