Artis Systems VD12: Post-processing Touchscreen Control Video
This video explains how to post-process 2D images using the buttons on the touchscreen control (TSC) and the mouse joystick.
Welcome to the Artis 2D postprocessing on the touch screen control TSC video. In this video you will learn how to post process 2D images using the buttons on your touchscreen control and the mouse joystick. With windowing you can adapt the image to display the grayscale region of interest with optimal contrast and brightness conditions. To window images, select the window button in the image task card. On the touch screen control TSC. Deflect the mouse joystick to adjust the window values to adjust the window center image brightness, deflect the joystick up and down. To adjust the window width, image contrast, deflect the joystick left and right. To reset the window values, press home with the left mouse button on the mouse joystick. To close windowing, press confirm with the right button on the mouse joystick. You can also adjust the edge enhancement, select the edge enhancement button. Deflect the mouse joystick up to apply more edge enhancement. Down to apply less edge enhancement. To reset the edge enhancement values, press on the left mouse button on the mouse joystick. To close edge enhancement, press on the right button on the mouse joystick. To set the electronic shutter, press on the shutter button in the image task card on the touch screen TSC, if configured in the layout, deflect the mouse joystick to adjust the shutter edges. Deflect the joystick to the right to close the left and right shutters. Deflect the joystick to the left to open the left and right shutters. Deflect the joystick down to close the upper and lower shutters. Deflect the joystick up to open the upper and lower shutters. Depending on the image matrix, the image type an the installed license is different. Zoom functions are available. Zoom by a factor of two and zoom to acquisition size. If configured in the layout, you can also invert the image from the touch screen control TSC. For scenes with an ECG, you can activate or deactivate the ECG display.
The reproduction, transmission or distribution of this training or its contents is not permitted without @ Siemens Healthcare GmbH, 2020 express written authority. Offenders will be liable for damages. Please note that the learning material is for training purposes only! All names and data of patients, parameters and configuration dependent designations are fictional and For the proper use of the software or hardware, please always use the Operator Manual or Instructions for examples only. n' 'Exam n' Quant' // Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be used Quanto 40 as training material only and shall by no means substitute the Operator Manual. Any material used in this All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Artis Systems training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. Adjust The Operator Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Post-processing Touchscreen Control Applications DSA Note: Some functions shown in this material are optional and might not be part of your system. REF Certain products, product related claims or functionalities described in the material (hereinafter collectively Tools "Functionality") may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information.
- artis q.zen
- artis qzen
- atris qzen