Artis zee multi-purpose VD12: System Movements Video
This video explains how to perform system movements on Artis zee multi-purpose using the control console.
Welcome to the system movements video for the artist. The multi purpose in this video you will learn how to perform system movements on Artis. See multipurpose using the control console. You use the stand control module SCM on the control console to control system movements. You activate system movements by deflecting the multifunctional joystick at the top of the multifunctional joystick is the dead man switch. You must press the Deadman switch prior to and during joystick deflection to activate system movement. The Deadman Switch helps protect against unintentional system movements that may occur as a result of accidental touch. Press down on the joystick and deflect the joystick in the desired direction. For example, Eleo and RA. Oh angulations. Press down on the joystick and deflect the joystick for cranial and caudal angulations. Please note the direction of joystick deflections for El, Ara oh and cranial caudal angulation depends on the location of the control modules in relation to the system. There is an orientation key on the stand control module to select the orientation. The illuminated LED indicates the active position. To change orientation, for example, when moving the control console to another location on the table, press an hold the key for more than three seconds, the system toggles to the next orientation. Keep pressing the key until the correct orientation is selected. There are buttons on either side of the multi functional joystick. When one of the buttons is pressed, an alternative see arm movement is activated. On Artis see multi purpose pressing. One of the side buttons activate stand lanja Tude Inal movements. Please note you only need to press one of the side buttons to activate the movement. You do not need to press both side buttons. On the back of the multi functional joystick there is a rocker switch for lifting and lowering the detector press on the lower button to lower the detector. Press on the top button to raise the detector. On the stand control module there are keys to select system positions and programmed positions. System positions are basic positions of the stand and table, for example, frontal, lateral over table and under table. Select the system positions key on the stand control module. A menu appears on the assist screen with the available positions. Deflect the joystick to select the desired position to confirm the selection, press one or both of the two buttons on the sides of the joystick. To move the system to the selected position, press down on the joystick and deflect the joystick away from you. The speed of system movements depends on the degree of joystick deflection. That is, maximum speed is achieved with maximum deflection. Keep the joystick deflected until the system reaches its position. When the position is reached, the lead on the key goes out. You can save a number of positions that you require for examinations under the programmed positions key. You can select the position and move to the position as described for the system positions. You can store your most frequently used positions as direct positions. There are three direct position keys on the stand control module. To store a direct position, move the system to the desired position. Press the store key. Press the desired position Key 1, two or three. The current position is stored under the direct position key. The previously stored position is overwritten. To move the system to a direct position, press the required position key, press down on the joystick and deflect the joystick away from you. The system moves to the position. You can use the zero Stop button to stop the see arm in the basic position. Press the Zero stop button. The LED lights up. Movement stop at the basic position.
The reproduction, transmission or distribution of this training or its contents is not permitted without @ Siemens Healthcare GmbH, 2020 express written authority. Offenders will be liable for damages. Please note that the learning material is for training purposes only! All names and data of patients, parameters and configuration dependent designations are fictional and examples only. For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be used too J S 90 as training material only and shall by no means substitute the Operator Manual. Any material used in this All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Artis zee multi-purpose training will not be updated on a regular basis and does not necessarily reflect the latest version of the SIEMENS software and hardware available at the time of the training. The Operator Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Healthineers System Movements Note: Some functions shown in this material are optional and might not be part of your system. eee Certain products, product related claims or functionalities described in the material (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information.
- artis. artsi