Atellica® Direct Load Monthly Maintenance PEP to Go Version 1.25 and Below
PEP to Go is an education resource material designed to support competency mastery on-the-job.
SIEMENS Healthineers 29 CH SIEMENS Healthineers Atellica® Direct Load Monthly Maintenance PEP to Go Effective Date: 5/21/2018 | HOOD05162002852946 Unrestricted © Siemens Healthcare GmbH, 2018 Atellica® Direct Load SIEMENS Monthly Maintenance Healthineers Monthly Maintenance Clean the DL fan filter 2 Effective Date: 5/21/2018 | HOOD05162002852946 Unrestricted © Siemens Healthcare GmbH, 2018 Atellica® Direct Load SIEMENS Monthly Maintenance Healthineers Clean the DL Fan Filter 1. Grasp the top of the fan filter and pull it up and out of the slot. The fan filter is located in the lower area of the back DL panel 2. Inspect and discard damaged filters 3. Rinse the filter with tap water 4. Tap the filter to remove excess water and let it air dry 5. Reinstall a spare fan filter with the plastic grid facing away from the module 6. If the original filter was discarded, order a replacement filter NOTE: The customer parts kit includes spare fan filters. Use a spare filter while the cleaned filter dries. When dry, place the cleaned filter in the customer parts kit for future use. 3 Effective Date: 5/21/2018 | HOOD05162002852946 Unrestricted © Siemens Healthcare GmbH, 2018 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2018 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 4 Effective Date: 5/21/2018 | HOOD05162002852946 Unrestricted © Siemens Healthcare GmbH, 2018