Atellica® IM Analyzer Monthly Maintenance PEP to Go Version 1.25 and Below
PEP to Go is an education resource material designed to support competency mastery on-the-job.
SIEMENS Healthineers Atellica® IM Analyzer Monthly Maintenance PEP to Go Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020 Atellica® IM Analyzer SIEMENS Monthly Maintenance Healthineers Monthly Maintenance ➢ Clean the Exterior of the Aspirate Probes ➢ Clean the IM Fan Filters 2 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020 Atellica® IM Analyzer SIEMENS Monthly Maintenance Healthineers Cleaning the Exterior of the Aspirate Probes SIEMENS Atellica Solution Version. 184.108.40.20683003 P-P1-04 A System Samples Inventory Calibration QC Patient Orders Worklist Maintenance Setup 1. Unlock the IM analyzer front cover: B Status Logs Operator Diagnostics About A. On the Command bar, select System System VMM P-P1-03 C-P1-06 >1-060 C -000 B. Select the Status tab Standby Standby Standby Standby C. Select the IM analyzer Capabilities Alert Category Time Luminometer V Reagent Loader D. Select to pause or stop the analyzer Reagent Compartment V Reagent Probe 1 E. Select Front Cover to unlock the front cover V Reagent Probe 2 Connected to PCC 2. Lift open the IM front cover D V Reagent Probe 3 V IM Acid II V IM Base Patient Sample Processing V IM Cleaner M Wash Front Cover V Special Reagent Water E Back Cover V Cuvette Loader Liquid Waste * This procedure is continued on the next slide. Tin Trav Waste ? PEP LabManager Wednesday, June 13, 2018 8:02 AM 3 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020 Atellica® IM Analyzer SIEMENS Monthly Maintenance Healthineers Cleaning the Exterior of the Aspirate Probes (continued) 3. Carefully lift the aspirate probe arms to the highest position Note: Do not remove the aspirate probes. Do not touch the probe tip when working with the probes. Do not bend the probe. 4. Clean the visible portion of each probe using a gentle downward motion with lint-free tissues moistened with IM Cleaner 5. Wipe each probe with lint-free tissue saturated with special reagent water, using a gentle downward motion 6. Repeat step five 3-5 times with clean lint-free tissues 7. Close the IM front cover 8. On the command bar, select System > Status, then select the IM analyzer 9. To restart the analyzer, select 10. Mark the Cleaning the Exterior of the Aspirate Probes activity as completed by selecting: Maintenance > Schedule > Cleaning the Exterior of the Aspirate Probes > Mark as Completed 4 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020 Atellica® IM Analyzer SIEMENS Monthly Maintenance Healthineers Cleaning the IM Fan Filter The IM analyzer has 1 filter located on the back center panel. 1. Grasp the left side of the fan filter and pull it out of the slot 2. Inspect and discard damaged filters 3. Rinse the filter with tap water 4. Tap the filter to remove excess water and let it air dry Note: The customer parts kit includes spare fan filters. Use a spare filter while the cleaned filter dries. Once dry, place the cleaned filter in the customer parts kit for future use. 5. Reinstall a spare fan filter with the plastic grid facing away from the panel 6. If the original filter was damaged or discarded, order a replacement filter 7. Mark the Cleaning the IM Fan Filter activity as completed by selecting: Maintenance > Schedule > Cleaning the IM Fan Filter> Mark as Completed 5 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 6 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020