Atellica® Sample Handler Connect Hardware Overview PEP to Go
PEP to Go is an education resource material designed to support competency mastery on-the-job.
SIEMENS Healthineers Atellica® Sample Handler Connect Hardware Overview PEP to Go 1 Effective Date: 2/20/2018 | HOOD05162002777696 Unrestricted © Siemens Healthcare GmbH, 2018 Atellica® Sample Handler Connect SIEMENS Hardware Overview Healthineers The Atellica Sample Handler Connect (SHC) is a sample transfer unit • Provides the hardware for transferring routine and STAT samples between a connected Lab Automation System (LAS) and the system The operator can simultaneously load and unload samples from the Sample Handler (SH) sample drawers while the system processes samples from the LAS • The SH can be used to load tube types the LAS does not support such as tube top cups or capillary tubes The SHC cover locks during normal operation • To open the cover, the operator requests to unlock the cover from the primary user interface 2 Effective Date: 2/20/2018 | HOOD05162002777696 Unrestricted © Siemens Healthcare GmbH, 2018 Atellica® Sample Handler Connect SIEMENS Hardware Overview Healthineers Automated robotic arms: A. Input robotic arm: Transfers sample containers onto Atellica Magline® B A Transport carriers on the SHC module B. Output robotic arm: Returns sample containers to the LAS Top down view of SHC TCS: SHC Tube Characterization System (TCS): • An integrated system of 3 cameras that reads the barcodes on the sample containers the SHC transfers from the LAS • The center camera also performs image analysis of the sample containers The cameras are in a triangular arrangement, which provides 360°imaging of • sample containers in the carrier • After the SHC TCS confirms the barcode matches the information from the LAS and determines the tube characteristics, the system routes the carrier to the appropriate analyzer NOTE: Do not position pacemakers or other implanted medical devices closer than 7.5 cm (3 inches) from the Atellica Magline® Transport. Carriers on the Atellica Magline Transport generate strong magnetic fields that can interfere with cardiac medical implants. 3 Effective Date: 2/20/2018 | HOOD05162002777696 Unrestricted © Siemens Healthcare GmbH, 2018 Atellica® Sample Handler Connect SIEMENS Hardware Overview Healthineers 6 Sample Processing Sequence Atellica Magline Transport carrier 1. The LAS provides the sample barcode ID to the system 2 2. The SHC input robotic arm transfers the sample from the LAS to a carrier on the Atellica Magline® Transport 3. The SHC tube characterization system checks the sample barcode to confirm it matches the LAS input • If the barcode information does not match, the system marks the sample as mismatched and the SHC output robot returns it to the LAS OOO • If the barcode information matches, the system identifies the required tests and continues processing the sample 4. The Atellica Magline Transport delivers the carrier and associated sample to the analyzer(s) for LAS carrier aspiration 5. After aspirating all tests, the system returns the sample to the SHC 6. The SHC output robotic arm transfers the sample to the LAS 4 IM Effective Date: 2/20/2018 | HOOD05162002777696 Hardware Overview Unrestricted © Siemens Healthcare GmbH, 2018 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2018 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 5 Effective Date: 2/20/2018 | HOOD05162002777696 Unrestricted © Siemens Healthcare GmbH, 2018