BioMatrix Head/Neck 20 (Numaris X)
This video explains the handling and features of the BioMatrix Head and Neck 20 coil neck for examinations of the head, neck and blood vessels
The Biomatrix head Neck 20 is used for examinations of the head and neck, such as the nape and blood vessels. An combined measurements with other coils. The coil contains the tilting mechanism to provide greater patient comfort, especially for examining patients suffering from special deformities. For example, Bekhterev's disease, a new feature of the coil is the integrated local B0 schimming, which helps to achieve higher image quality for specific parts of the anatomy. For neck region examinations or for cervical spine examination, the upper part of the coil can be omitted for slender patients. To use the coil, slide the Biomatrix head neck 20 into the lateral recesses at the head end of the patient table so that it locks into coil sockets five and six. You will see that the two parts of the coil are correctly plugged in on the PD. Move the lever to release the upper part of the coil and lift it up and off. Place the pad in the lower part of the coil. Position the head of the patient in the lower part of the coil. Ensure that the shoulders touched the lower part of the coil. Tilt the coil into the required position where necessary. For tilting and tilting back again, release the lever on the left or right side of the coil and at the same time lift the coil into the tilt position. If you can only lift the coil with one hand, grip the lateral recess. Pull the lever with your thumb and tilt the coil. Alternatively, tilt the coil with both hands using one hand to release the lever and using the other hand to grip the recess on the back of the coil to tilt it. If the coil is already in a tilted position, make sure that you do not accidentally release the lever. Always hold the coil before you release the lever to prevent it falling back into the O position. Use suitable positioning aids to improve patient comfort. Secure the patient's head using the corresponding cushions. Place the upper part of the lower part and slide it forward until it locks into place. You will see that the two parts of the coils are now correctly plugged in and registered on the PD. A mirror can be attached to the upper part of the coil to improve patient comfort during
upper part,lower part,lateral recess,coil plugged,position,patient,parts,lift,release,tilt,coil,coils,slide,locks,lever,place,hand,correctly plugged,operator manual,hereinafter collectively,regular basis,necessarily reflect,latest version,siemens •,training,siemens,material,head,healthineers,neck
SIEMENS .. Healthineers SIEMENS .. SIEMENS Healthineers SIEMENS • SIEMENS .. Healthineers SIEMENS • SIEMENS • SIEMENS Healthineers SIEMENS ... SIEMENS • SIEMENS .. SIEMENS Healthineers Healthineers SIEMENS •. Healthineers SIEMENS ... SIEMENS No patient registered Bram O Siemens Healthcare GmbH, 2018 Bramn Healthineers MAGNETOM systems Brain Please note that the learning material is for training purposes only! Tablo Position Home Head BioMatrix SIEMENS .. SIEMENS Home SIEMENS C-Spine I Neck For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use BM 20 Healthineers spine 32 (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be used as BM spine 32 Coils Healthineers BM Head/Neck 20 BM 20 training material only and shall by no means substitute the Operator Manual. Any material used in this training L-Spine BM spine 32 post 0 deg T-Spine will not be updated on a regular basis and does not necessarily reflect the Latest version of the software and will not be updated on a regular basis and does not necessarily reflect the latest version of the software and 00 Direct BM Head/Neck 20 00 hardware available at the time of the training. post 0 deg ant BM 20 Connect ant COIL PLUGGED The Operator Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. COIL PLUGGED Note: Some functions shown in this material are optional and might not be part of your system. The information MAGNETOM BM Head/Neck 20 ant BM 20 ant in this material contains general technical descriptions of specifications and options as well as standard and The reproduction, transmission or distribution of this training or its contents is not permitted without express optional features that do not always have to be present in individual cases. written authority. Offenders will be liable for damages. BM Head/Neck 20 ant A BioMatrix System Certain products, product related claims of functionalities described in the material (hereinafter collectively ALL names and data of patients, parameters and configuration dependent designations are fictional and 00 "Functionality") may not (yet) be commercially available in your country. Due to regulatory requirements, the examples only. future availability of said Functionalities in any specific country is not guaranteed. Plaese contact your local Siemens Healthineers sales representative for the most current information. All rights, including rights created by patent grant or registration of a utility model or design, are reserved.