C0 Adjust Pack Calibrations on the Atellica® CH Analyzer Job Aid Version 1.25 and Below
This job aid provides an overview of processing C0 adjust pack calibrations on the Atellica® CH Analyzer.
Atellica® CH Analyzer SIEMENS C0 Adjust Pack Calibrations Job Aid Healthineers C0 Adjust Pack Calibrations The CH Analyzer supports 2 types of pack calibrations: Full – multiple calibrator levels are used C0 Adjust – only a low level calibrator is used o For Siemens assays that use C0 Adjust pack calibrations, CH Diluent is used as the calibrator o For open channel assays that use C0 Adjust pack calibrations, either a zero level calibrator or CH Diluent is used as the calibrator The CH assay test definition specifies the type of pack calibration an assay requires. Only certain assays use C0 Adjust pack calibrations. For assays that use C0 Adjust pack calibrations: When the system alerts that a pack calibration is needed because the previous pack calibration has expired, and the operator orders a pack calibration, the system will perform a C0 Adjust pack calibration. For Siemens assays, the operator does not need to load calibrator samples in this case because the system automatically uses CH Diluent When the system alerts that a pack calibration is needed because the previous lot calibration has expired, and the operator orders a pack calibration, the system will perform a full pack calibration. The operator must load the appropriate calibrator level samples 1 Effective Date: 11/12/2018 | HOOD05162002965558 Unrestricted © Siemens Healthcare GmbH, 2018 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability may vary from country to country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2018 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 2 Effective Date: 11/12/2018 | HOOD05162002965558 Unrestricted © Siemens Healthcare GmbH, 2018