Changing the IMT Peristaltic Tubing on the Atellica® CH Analyzer Video Version 1.25 and Below
This video provides instructions on how to perform the as needed maintenance task of changing the IMT peristaltic tubing on the Atellica® CH Analyzer.
will show you the steps on how to change the empty peristaltic tubing on the Atellica chemistry analyzer. The analyzer. will post an alert to replace the empty peristaltic tubing on a yearly basis. To change the tubing, the system must be set to diagnostic state. This can be done by selecting system in the command bar. Next, select operator diagnostics. Select the chemistry module icon. Select enter diagnostics. Select yes in subsystems select IMT. Select replacing the IMT peristaltic pump tubing. Select perform select Yes, wait for the analyzer to prepare for maintenance. Next, open the front cover of the analyzer. The pump is located to the left of the lamp coolant container behind the IMT fluids. Lift the peristaltic pump cover and pull out the tubing from the V channel. Unscrew the tubing from both ends and discard the tubing per laboratory protocol. Attached the new tubing by screwing in each end one at a time. When secure the nob clicks. Place the tubing in the V channel. Close the pump cover. And close the analyzer front cover. To exit diagnostics, select continue in the software. Next, select close. Select exit diagnostics. Finally, select yes. To complete the procedure, select the maintenance tab in the Command bar, select schedule. So like the chemistry module from the dropdown menu. Select replacing the IMT peristaltic pump tubing in the software mark I empty peristaltic pump tubing replacement as completed. After the tubing is changed perform and IMT calibration manually to complete the procedure to perform and I empty calibration. Select the calibration tab in the software. Next, select EMT calibration. Finally select calibrate EMT.
command bar,front cover,select continue,select,close,software,change,analyzer,channel,exit diagnostics,washer probes,set warning,add activty,operator diagnostics,ch frequency,ch maintenance,select replacing,perform replacing,maintenance,pump,replacing,enter,perform,system,time,module,scheduled,procedure,completed,tubing
Atellicae Solution Version 126.96.36.19983003 Atellicae Solution Version I Atellicae Solution Version I.14.0.2983003 SIEMENS SIEMENS . Atellicae Solution Version I .14.0.2983003 SP00107 SP00107• .14.0.2983003 Healthineers Healthnineers SIEMENS ... SIEMENS .. 11 Healthineers System Subsystems Cümion M airtenarre Seno Setto Inveitory ln•æltory Inventory Intæltory Invaltory Invertory Inve•tory Orders Or&ers Or±ers Or&rs Worklist Workhst Workhist M airtenaree M airtenare:e M aimtenarx:e M aiNenarx:e CH Maintenance Maintenance Report Data Restore Software Updates Copy Configuration Exit Diagnostics Operator Diagnostics Calibrator Definitions IMT SH SH 11 Module. CH Frequency: Al Tune: All Performed By: All Module: CH Frequency Al Time: All Performed All Module: CH Frequency: Al Time: All Performed All Filter CH SH System CH Module Activity Scheduled Fri-23 Sat-24 Sun-25 Sun-25 Thue-27 Mon-26 Tue-27 Thue-29 Wed-28 Thu-29 Calibration Subsystems Diagnostic Name Diagnosuic Name Diagnosuc Name Description Descnptton Descnptjon Ail All STD A STD B Ion Slope Frequency ATELLICA.CH Inspecting the Washer Probes Clspeaning the Washer Probes 1104/2018 5:00 Mark as Completed as Completed Confirmation O n f i IMT Penstalt•c Pump Tub•ong O n f i Peristaltic Pump Tubing Standby Standby Stand Repoacing the IMT Peristaltic Pump Tubing Replacing the IMT Peristaltic Pump Tubing Replacing the I MT Peristaltic Pumnp Tubing ' Replacing the I MT Peristaltic Pumnp Tubing the I MT Peristaltic Pump Tubng Pump Tubing Replacing IMT Peristaltic Pumnp Tubing Stand Replacing the IMT Peristaltic Purnp Tubing Replacing the IMT Peristaltic Purnp Tubing Lamp Perform Time 5902 Replacing the IMT Peristaltic Pump Tubing: Replacing the 1M T Peristaltic Pum -144271 mV Tubing SO t063 50t063 CH CH Weekty Maintenance CH Maintenance M airtenarr:e CH Weekly Maintenance CH l•ßaintenance 11/11/2018 4:00 11/04/2018 5:00 Not Performed. Category Categooy' Invertory Time Confirmation All Performed By K. -1535 50t063 Capabilities Checking the Larnp Coolant Checking the Lamnp Coolant 6:00 Procedure Details Procedure IMT Replacing the IMT Peristaltic Pump Tubing Changing the IMT Peristaltic Tubing Enter the actMvity name Enter the activity name Search Text 8308 CH Maintenance CH M•ßaintenance CH Dai}! Maintenance 4:00 Scheduled by Operator CH Mahtenance CH Maintenancq 2 V 01/01/2019000 Perform: Replacing the IMT PeristaRic Pump Tubing? Perform Replacing the IMT Peristaltic Pumnp Tubing? Perform Replacing the IMT PeristaRic Pump Tubirg? I. Refer to the online help for the procedure: Replacing the IMT Completing Replacing the IMIT Peristaltic Pump Tubing: Replacinghe IMT Peristaltic Pump Tubing: Peristaltic Pump Tubing. Scheduled Automatically ATE-LUC-A.CH Replacing the IMT Peristaltic Pump Tubing Replacing IMT Peristaltic Pump Tubing 11/252020 15:48 I-quid Reading Liquid Reading 0234 v Autocheck Scheduled by Serve Scheduled by Service 2. After completing the procedure, select Continue. Cleaning CH Cwettes Cleaning CH Ccrvettes Connected to PCC MT Rate IMT Rate Failted Failter Faded ATE-LUICA.CH Cleaning Reagent Loader Overdue IMT Calibration Completed IMT Calibraton Completed Calibration Completed Inspect and Clean Inspect ard Clean Inspect Clean axe Inprocess Unlock Lids Continue Cancel Close Cleaning System Fluids Completed as Completed Clearing Washer Probes Front Cover Not Perform.. Not Performed. Not Performed Draining and Filling the Reaction Ring Bath Back Cover With Comments Refreshing Water Bath Additive Refreshing the Water Bath Additive As Needed As Neeckd ATELLIC-ACH ATELLIC-A.CH ATELUICACH Replacing Reagent Mixer Irnpener Replacing Reagent Mixer Irnpelkr As Needed Needed Enter Diagnostics Exiter Diagnostics Operator Diagnostics Eniter Diagnostics Operator Diagnostics Exiter Diagnostics Exit Diagnostics Set Warning... Add Activty... Add Activty.. Set Warning.. axe PEP LabM anager Tuesday. November 27, 2018 25129 PM Monday, November 26, 2018 32910 PM Tuesday. 27, 2018 25129 PM Monday, November 26, 2018 348:35 PM Monday, November 26, 2018 3:4835 PM Tuesday, July 10, 2018 AM LabManager LalbManager Monday, November 26, 2018 34835 PM Monday, November 26.2018 PM Monday, November 26, 2018 316:33 PM Monday, November 26, 2018 3$33 PM Monday, November 26. 2018 34835 PM Tuesday. November 27, 2018 PM Monday, November 26, 2018 329103 PM Monday, November 26. 2018 PM