Checking and Filling the Lamp Coolant on the Atellica® CH Analyzer Video Version 1.25 and Below
This video provides instructions on how to perform the weekly maintenance task of checking and filling the lamp coolant on the Atellica® CH Analyzer.
This video will show you the steps on how to check and refill the lab coolant on the atellica solution Ch Analyzer. The Ch lamp coolant must be checked and refilled once a week to access the lamp coolant reservoir. The system must be set to diagnostic state. This can be done by selecting system on the command bar. Next, select operator diagnostics. Select the Ch module. Select enter diagnostics. Select yes in subsystems, select lamp. Select checking the lamp coolant. Select perform. Select yes, wait for the analyzer to prepare for maintenance. Open the front cover of the analyzer. The cover automatically unlocks after the analyzer has prepared for maintenance. Check the level of the coolant in the reservoir by viewing from the right side. If the coolant is not visible, it needs to be refilled. The box for the lamp coolant contains a bottle of lamp coolant and a spigot to allow for easier pouring. Remove the cap from the lamp, coolant bottle and attached the spigot. Unscrew the small screw cap on the top of the lap coolant reservoir and set aside. Using the spout on the lab coolant bottle, squeeze coolant into the reservoir until the fluid level is approximately 1 to 2 centimeters from the lab coolant reservoir cover. Once you have refilled the reservoir with enough coolant, replace the small screw cap on the reservoir and tighten until secure. Close the cover to the chemistry analyzer. To exit diagnostics, select continue in the software. Next select close. Select exit diagnostics. Finally, select yes. To complete the procedure, select the maintenance tab in the command bar. Select schedule. Select the chemistry module from the dropdown menu. Select checking the lab coolant in the software. Finally, mark checking the lamp coolant as completed.
command bar,select checking,coolant,select,analyzer,close,software,exit diagnostics,source larrv,aspirate probes,reagent probes,front cover,frck cover,lab coolant,amp coolant,source lamp,product availability,lamp coolant,operator diagnostics,cleaning extenor,lamp,operator,checking,extenor,maintenance,database,scheduled,module,healthineers,cleaning
Atellicae Solution Version 18.104.22.16883003 Atellicae Solution Version 1 .14.0.2983003 Atellicae Solution Version I SIEMENS . SIEMENS SIEMENS .. sp00107• SP00107 SP00107• Henlthinpprs Healthineers Heotthineers SIEMENS .. SIEMENS d Ill] 11 Healthineers Healthineers • System Subsystems Workli< Workw M antenarx:e M aiMenaree Sento Setto In valtory InveNtory Invertory Invaltory Inve-dory Inveütory C&æion Orders Or&ers Ordets Workhst M aimenaree Report Data Restore Software Updates Copy C onfiguration Copy Configuration Status Operator Diagnostics Exiter Diagnostics Operator Please note that the learning material is for training purposes only! CH SH 1M Filter Module; A." Frequency All Time AJI Performed All Module: All Frequency: All Time: All Performed By: All ATELLICA ATELLICA.IM ATELLICACH System VMM Module Mohedule Activity ActiviW Scheduled Fri-22 Sat-23 Sun-24 Mon-25 sun-24 Wed-27 Thu-28 Fri-a Sat-as Subsystems Diagnostic Name Description All AT ELLICNCH Inspecting and Cleaning the Probe Washr Stations Inspecting and Cleaning the CH Probes and Mixer Impellers Inspecting and Cleaning the CH Probes ard Mixer Impellers 0301/2019 Mark as Cornpleted Standby Standby Replacing the Source Lamp Replacing the Source Lampv Replacing Source Lamp Replacing the Source Larrv Replacin"g the Source Lamp Replacing the Source Larrv Replacing the Source Lamnp Lamp Perform All Checking Lamp Coolant (D CH Time CH Checking Lamnp Coolant AT ELLICA.CH Inspecting and Cleaning Probe Wash Stations Inspecting and Cleanyng the Probe Wash Stations Not Checking Lamp Coolant Checking the Lamp Coolant L amp Coolant Checking Lamp Coolant L amp Coolant Lamp Coolant Lamp InveNtory Category Tme Confirmation All Capabilities Cleaning the CH Fan Filter Cleaning 1M Fan Filter Cleaning CH Fan Filter Procedure Details Enter the act.wty name Enter the activity name Search Text Cleaning Extenor of the Aspirate Probes Cleaning Extenor the Aspirate Probes 032512019 5:30 Cleaning the Extenor of the Aspirate Probes Check and Replace the Lamp Coolant: EdWVæw.. EdWVæw... ATELUCA-IM Scheduled by Operator 7:00 Replacing the Source Larrv Pedformj Replacing the Source Lamp? Refilling the CH Lamp Fluid Scheduled Automatical>yr Scheduled Automatically Scheduled Automaticallyr Cleaning Extenor the Reagent Probes Cleaning the Extenor of the Reagent Probes Cleaning Extenor of the Aspirate Probes Cleaning the Extenor of Reagent Probes CH Autocheck Auto-check Scheduled by Servxe Scheduled by Service Inspecting and Ernptying the 1M Water Trap Inspecting and Cleaning the Probe Washr Stations Inspecting and Ernptying the' 1M Water Trap 0301/2019 0:00 0301/2019 03021/2019 5:00 Connected to PCC Not Yes No 1M WeekFy' Maintenance 03021/2019 7:00 Overdue Inspect ard Clean Inspect Clean lnspect and Clean Inprocess ATELLICA.CH ATELLICN.CHM 1M WeekFy' Maintenance CH Weeny' Maintenance CH Maintenarxe CH Weekty Maintenarxe CH Weeny Maintename CH Maintenance 0324/2019 400 Close Unlock Covers Continue Frck Cover SIEMENS Product availability may vary from country to country and is subject to varying regulatory requirements. Please contact y Product availability may vary from country to country and is subject to varying regulatory requirements. Please contact your local representative for availability. Completed ATELLICA-CH Checking the Lamp Coolant Checking Lamp Coolant Front Cover Frck Cover Not Performed Database kup Database Backup Back Cover Frck Cover With Comments Database: Archiving and Database: Archiving and Deleting 0&25.'2019 3:00 0325.12019 3:00 300 7:00 6:00 03202019 300 Database: Database • Optimnizing Database • Optimizing 032512019 5:30 5:30 Database OptJrniang 0325/2019 5:30 0301/2019 7:00 Exiter Diagnostics Enter Operator Diagnostics Eniter Diagnostics Operator Diagnostics Exit Diagnostjcs Enter Diagnostics Exit Diagnostics Exiter Diagnostics Add Set Warnim Set Warnirg PEP LabM anager Monday, March 25.2019 AM Monday, March 25, 2019 AM Thursday, 14.2019 11:19:57 AM Thursday, 14.2019 11:19:57 Tuesday, July 10, 2018 112730 AM LabManager Thursday, 14, 2019 AM Thursday, February 14, 2019 AM Thursday, February 14, 201911B009AM Thursday. 14, 2019 AM Thursday, 14, 2019