Cleaning Sample Spills on the Atellica Magline® Transport Video Version 1.25 and Below
This video will explain how to clean sample spills on the Atellica Magline® Transport.
This procedure can be used to clean a spill that occurs within the atellica magline transport. 1st put the Atellica magline transport into stop mode by selecting the system button on the command bar. And then selecting the Status tab and then the Atellica magline transport module icon on the Atellica magline Transport status screen. Select the stop icon. Remove one or more atellica magline transport covers to access the spill area by first locating the screws under the Atellica magline transport cover. Then loosen the two atellica magline transport cover screws enough to allow you to slide the cover off. You do not need to completely remove the cover screws. All the atellica magline transport cover away from the Atellica Magline transport and set aside. Clean the spill with lint free tissues or gauze saturated with a 10% bleach solution and finally drying the area thoroughly with clean lint, free tissues or gauze. Then reinstall the Atellica magline transport covers by positioning each cover so the edges fit into the slots on either side of the opening. Slide the Atellica magline transport cover in until the spring clips lock the cover into place. Tighten the two screws under the Atellica magline transport cover. To complete the procedure, start the Atellica magline transport by selecting the resume icon on the Atellica magline Transport System status screen.
free tissues,cover screws,spill,remove,area,transport,clean,screws,atellica,cover,procedure,hereinafter collectively,means substitute,regulatory requirements,future availability,operator manual,siemens healthineers,specific country,training material,manual,operator,availability,diagnostics,country,siemens,names,material,training,system,local
Atellica• Solution Calbation Samples Inventory Calbation PatieN Patient Orders Setup Please note that the learning material is for training purposes only! operator Operator Diagnostics For the proper use Of the software, please always use the Operator Manual Or Instructions for Use (hereinafter collectively "Operator Manual") Diagnostics issued by Siemens Healthineers. This material is to be used training material only and shall by no means substitute the Operator Manual. issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software System .01 CH and hardware available at the time Of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Standby Note: Some functions shown in this material are optional and might not be part of your system. Diagram Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. Capabilities Due to regulatory requirements, the future availability Of said Functionalities in any specific country is not guaranteed. Please contact your local Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local SIEMENS Siemens Healthineers sales representative for the most current information. Atellica W i n e Transport Atellica Transport The reproduction, transmission or distribution Of this training or its contents is not permitted without express written authority. Offenders will be Liable for damages. The reproduction, transmission or distribution Of this training or its contents is not without express written authority. Offenders will be Liable for damages. C. p I .01 AIL names and data Of patients, parameters and configuration dependent designations are fictional and examples only. All names and data Of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. SK-PI -01 c-p1-01 connected to pcc Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. please contact your local representative for availability. Copyright O Siemens Healthcare GmbH 2020 Global Siemens Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 913184 0 siemens.com/healthineers Tuesday, 2B, Tuesday, February 2B, 2017 102234 AM Tuesday, 2B, 102352 AM