Cleaning the Atellica® IM Analyzer Strap Sensor Video Version 1.25 and Below
This video will explain how to clean the Atellica® IM Analyzer strap sensor.
This video will show the procedure that can be used to clean the strap sensor on the atellica. I am analyzer. First ensure that the I am analyzer is in the stopped state by selecting the system button on the command bar. Next select status. Then select the atellica. I am analyzer icon. If the analyzer is not already stopped, select the pause button to allow the system to complete any testing currently in process. The system will pause and then go to stop state. Lift up and open the IM Tiptree loader cover. Remove any obstructions from the strap sensor using a lint free swab or tissue dampened with alcohol. Insert into the recessed sensor area and wipe the lens three times when complete. Be sure to remove the cotton swab from the sensor area. Close the IM Tiptree loader cover. To complete this procedure, resume the IM analyzer by selecting the resume button on the IM System status screen to bring the analyzer back to the ready state. This completes the procedure to clean the strap sensor on the atellica I am analyzer.
strap sensor,sensor area,iam analyzer,analyzer,stopped,pause,resume,select,system,hereinafter collectively,main reference,ent water,regulatory requirements,local representative,operator manual,specific country,siemens healthineers,manual,operator,water,local,requirements,patients,healthineers,material,country,patent,training,siemens,availability
M *tenure Please note that the learning material is for training purposes only! Operator Di•ønostics For the proper use Of the software, please always use the Operator Manual Or Instructions for Use (hereinafter collectively "Operator Manual") System issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. VMM -03 D-pp-15 Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time Of the trainingy and hardware available at the time Of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. The Operator's Monunl shall be used os your main reference. in pnrticular for relevant sntety information like warnings and cautions. Standby Cleaning the [email protected] Note: Some functions shown in this material are optional and might not be part of your system. Capabilities Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. SIEMENS Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your Local Due to requirements, the future smd any specific country is not guaranteetl Please contact your Local Siemens Healthineers sales representative for the most current information. Sientens Healthineers smes representative for the 'nost current 1M Analyzer Strap Sensor Video Cornpattrnent The reproduction, transmission or distribution Of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. The reproduction, transmission or distribution of this training or its contents is not pernlitted without express written authority. Offenders will he liable tor damages. All names and data patients, paramleters and configuration dependent designations are fictional and examples only. ALI nomes and dato of patients, parameters and configuration dependent designations are fictional and examples only. Healthineers All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Connected to PCC Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands ore the property of their respective owners. Base 11 Product availability varies by country and is subject ta varying regulatory requirements. Please contact your local representative for availability. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Base Cleaner Copyright O Siemens Healthcare GmbH 2020 Patent Prcressim Patent Rement Water Water Patent Watert Global Siemens Healthineers Headquarters Frcnt Cover Henkestr. 127 •V Special Re»ent Water Cwene 91052 Erlangen, Germany Back Cover Phone: +49 913184 0 Phone: +49 9131 84 0 siemnens.comn/heolthineers siemens.com/heolthineers Tray Waste Patent 755: 57 7:57 T 57 755 (57 LabManag•r