Cleaning the Atellica® Direct Load Entry and Exit Queues Job Aid Version 1.25 and Below
This job aid provides instructions on how to perform the as needed maintenance task of cleaning the Atellica® Direct Load entry and exit queues.
Atellica® Direct Load SIEMENS Clean the DL Entry and Exit Queues Job Aid Healthineers Clean the DL Entry and Exit Queues SIEMENS Build 1.8.0 160 K 1. Remove debris from the DL entry and exit queues System Samples Inventory Calibration QC About 2. Pause the DL: Status Logs Operator Diagnostics System DL-P1-01 C-P1-03 A. Navigate to the DL Status screen: System > Status > DL icon Ready Stopped B. Select the Pause icon Alerts Capabilities Alert 3. Wipe the lanes with lint-free tissue dampened with a V IPQ Processing 10% bleach solution V Rack Processing Routine Input V Rack Processing Routine Output 4. Wipe the lanes with lint-free tissue dampened with V Module Stopped reagent water Connected +- DCC B 6 5. Dry the lanes thoroughly with clean lint-free tissue 6. Start the DL by selecting the Resume icon on the DL Front Cover Status screen Back Cover HOOD05162002852946 Effective date: 5/21/2018 1 Unrestricted © Siemens Healthcare GmbH, 2018 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH 2018 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers HOOD05162002852946 Effective date: 5/21/2018 2 Unrestricted © Siemens Healthcare GmbH, 2018