Cleaning the Atellica® Sample Handler Sample Drawers and Racks Job Aid Version 1.25 and Below
This job aid provides instructions on how to perform the as needed maintenance task of cleaning the Atellica® Sample Handler sample drawers and racks.
Atellica® Sample Handler SIEMENS Clean the SH Sample Drawers and Racks Job Aid Healthineers SIEMENS Build 8.6.1 SH-P1-01 SH-P1-01 Cleaning the SH sample drawers and racks is an as needed maintenance task: 160 K 19 1. Stop the SH: System Samples Inviy Qc Patient Orders Worklist Maintenance Setup Status Logs CC/DL Operator Diagnostics Operator Diagnostics About PEPconnect A. Navigate to the SH Status screen: System > Status > SH icon System VMM SH-P1-01 CH IA-03 B. Select the Stop button 2. Open all drawers Standby Standby Standby Stopped 3. Unload all racks from the SH drawers and remove all sample containers from the racks Alert Category Time Capabilities 4. Clean the interior of each drawer: V Robot Wipe the area with lint-free tissues or gauze saturated with a 10% bleach solution V TCS A. Cal-QC Storage B. Wipe the area with lint-free tissues or gauze saturated with water Cal-QC Storage Temperature 1 d to Po 7 Drawer 1 C. Dry the area thoroughly with clean lint-free tissues or gauze Drawer 2 5. If the rack is contaminated or has stuck-on debris: V Drawer 3 A. Soak the rack in a 10% bleach solution for several minutes until the contamination Unlock Front Cover V Drawer 4 ¥ Front Cover or debris is softened or removed Jnlock Back Cover ¥ Back Cover B. Rinse the rack thoroughly with water C. Dry the rack thoroughly with clean lint-free tissues or gauze 6. If the rack requires only routine cleaning: ? LabManager2 Tuesday, February 28, 2017 10:14:41 AM A. Wipe the rack with lint-free tissues or gauze saturated with a 10% bleach solution B. Wipe the rack with lint-free tissues or gauze saturated with water C. Dry the rack thoroughly with clean lint-free tissues or gauze 7. Start the SH by selecting the Resume icon on the SH System Status screen HOOD05162002822815 Effective Date: 3/12/2018 Page 1