Cleaning the Sample Tip Drip Tray on the Atellica® IM Analyzer Video Version 1.25 and Below
This video describes how to perform the weekly maintenance task of inspecting clean the sample tip drip tray on the Atellica® IM Analyzer.
The sample tip drip tray on the atellica I am analyzer must be inspected weekly and cleaned as needed to prevent dry serum buildup and possible improper rejection of sample tips. This video will show the procedure that can be used to inspect and clean the sample tip drip tray on the atellica. I am analyzer first but the I am analyzer into the stopped state by selecting the system button on the command bar. Next select status. Then select the atellica. I am analyzer icon. If the analyzer is not already stopped, select the pause button to allow the system to complete any testing. Currently in process, the system will pause and then go to stop state. Then unlock the. I am back cover. And open the I am back cover. You must open the cover within 30 seconds of unlocking, otherwise the cover will relock while performing this procedure, ensure that loose debris does not fall into the incubation ring. There is an opening to the incubation ring near the sample tip drip tray. Remove the two thumb screws securing the drip tray and set aside. Remove the sample tip drip tray lifting up and to the left. Remove and discard any loose debris to clean the sample tip drip tray. First soak it in 10% bleach solution for five minutes. Then remove any remaining debris with a damp sponge or paper towel. Ensure there is no residue inside the drip tray and then rinse the tray with tap water. Complete the cleaning of the drip tray by drying the tray using a lint free tissue. After cleaning, install the sample tip drip tray. Ensure the tray is seated properly. Tighten the two thumb screws to secure the drip tray. Close the IM back cover. Enter I am diagnostic state my selecting system in the command bar. Then select operator diagnostics. Select the IM analyzer. And select enter diagnostic state. Then, under subsystems, select incubation ring. Select empty and fill ring. Select perform and then yes. When the procedure completes, select exit diagnostic state to complete this procedure, resume the IM analyzer by selecting the resume button on the IM System status screen to bring the analyzer back to the ready state. This completes the procedure to clean the sample tip drip tray on the atellica. I am analyzer.
System M enarre Please note that the learning material is for training purposes only! Operator Di•ønostics Operator Diagnostics For the proper use Of the software, please always use the Operator Manual Or Instructions for Use (hereinafter collectively "Operator Manual") -03 System VMM issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. IA-Ol Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time Of the training Diagnosbc Diagnosbics The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Standby Standby Errvty all the cuvettes fromn the incubation Empty a"l the Cuvettes frorn the irrubation Empty Ring Empty a" Cuvettes frorn the irrubatx)n Note: Some functions shown in this material are optional and might not be part of your system. Cleaning the [email protected] 1M Analyzer empty the cuvettes the the ring with new cuvettes, Empty and Fill Empty Ring Empty and Empty and Fil Capabilities SimultanecxJsty empty the cuvettes the and bad the runog with cuvettesv Simultaneously empty the cuvettes the ncubation rirO aryl bad the nng w•ith rww cuvettes. Simultaneously empty the cuvettes the ncubatjon and bad the nng with new cuvettesv empty the cuvettes the and bad the ring cuvettes Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. Rea;ent t aader SIEMENS Ring Initialize Move the subsystem to prünary Move the subsystem to prinary Re»ent Water Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. Sample Tip Drip Trag Video Cornpattrnent The reproduction, transmission or distribution Of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data patients, parameters and configuration dependent designations are fictional and examples only. Empty Empty Ring Empty all the cuvettes from the incubation All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Connected to PCC Reagent Empty and Ring Simultaneously empty the Cuvettes from the ring load nng with new Cuvettes Simultaneous"/ empty the cavettos from the Incubation ring and bad the ring with now cuvettes Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. B•aste Wash Reagent 11 Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Base Initialize Cleaner Copyright O Siemens Healthcare GmbH 2020 Patent Prcressim Reagent Water Patent Global Siemens Healthineers Headquarters Base B•aste Frcnt Cover F rmt Cover FrmtCover Cover Henkestr. 127 Status •V Special Reagent Water Cwene Re»ent Water 91052 Erlangen, Germany Cwene Phone: +49 913184 0 siemens.com/healthineers Tray Waste Water Exit State Status Turn Mechanks Off Turn Of Turn On 7:5767 7:57 : 57 757 (57 75557 757 T 57 57 57 LabManag•r Wednesday. 18.2020 AM Wedresday. 18.2020 18.2020