Cleaning the System Fluid Drawer on the Atellica® IM Analyzer Job Aid Version 1.25 and Below
This job aid provides instructions on how to perform the as needed maintenance task of cleaning the system fluid drawer on the Atellica® IM Analyzer.
Atellica® IM Analyzer SIEMENS Cleaning System Fluid Drawer Job Aid Healthineers The operator must clean the system fluid drawer as needed to ensure it is free from dust, dirt, or liquids. 2. Tighten the cap on each of the system fluid containers 1. Locate and open the system fluid drawer 3. Remove the system fluid containers * This procedure is continued on the next slide. 1 2 O O 1 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020 Atellica® IM Analyzer SIEMENS Cleaning System Fluid Drawer Job Aid (continued) Healthineers Note: Do not touch the cannulas in the keyed areas at the bottom of the system fluid drawer. 4 4. Carefully wipe the system fluid drawer with lint-free tissues or gauze saturated with special reagent water 5. Replace the system fluid containers O 6. Open the system fluid container cap with a one-quarter turn 7. Close the system fluid drawer 8. Mark the Clean System Fluids Drawer activity as completed by selecting: Maintenance > Schedule > Clean System Fluids Drawer > Mark as Completed Note: The analyzer logs the activity as completed in Maintenance > Log. 2 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 3 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020