Cleaning the Waste Area on the Atellica® IM Analyzer Job Aid Version 1.25 and Below
This job aid provides instructions on how to perform the as needed maintenance task of cleaning the waste area on the Atellica® IM Analyzer.
Atellica® IM Analyzer SIEMENS Cleaning Waste Area Job Aid Healthineers Cleaning the IM Waste Area involves cleaning the Sample Tip Waste Area and cleaning the Cuvette Waste Area. 1. Ensure the analyzer is in the paused or stopped state 2 A. On the Command bar, select System B. Select the Status tab 3 3 C. Select the IM analyzer D. Select to pause or stop the analyzer 2. Open the solid waste door 3. Pull out the solid waste drawer * This procedure is continued on the next slide. 1 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020 Atellica® IM Analyzer SIEMENS Cleaning Waste Area Job Aid (continued) Healthineers 4. Clean the waste bins: A. Remove the sample tip waste and cuvette bin liners B. Dispose of the sample tip waste in accordance with laboratory protocols C. Clean the waste bin liners using 10% bleach solution D. Rinse the bin liners with tap water E. Dry the bin liners with lint-free tissue F. Install the bin liners 5. Push the waste drawer in 6. Close the solid waste door 4 7. On the command bar, select System > Status, then select the IM analyzer 8. To restart the analyzer, select 9. Mark the Cleaning the Sample Tip Waste Area or Cleaning the Cuvette Waste Area activity as completed by selecting: Maintenance > Schedule > Cleaning the Sample Tip Waste Area or Cleaning the Cuvette Waste Area > Mark as Completed 2 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 3 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020