Clearing the IM Cuvette Loading Area Job Aid Version 1.25 and Below
This job aid describes how to clear the cuvette loading area on the Atellica® IM Analyzer.
Atellica® IM Analyzer SIEMENS Healthineers Clearing the IM Cuvette Loading Area Job Aid A System Samples Inventory Calibration QC Patient Orders Worklist Status Logs Operator Diagnostics About 1. Unlock and open the IM Analyzer front cover: System VMM P-P1-03 D-PP-15 C-P1-06 A. On the command bar, select System > Status B P1-06 B. Select the IM Analyzer Standby Standby Standby Standby Stopped C. Ensure the analyzer is in the pause or stopped state by selecting the pause or stop icon Capabilities Alert - Category V Reagent Loader D. To unlock the cover, select Front Cover V. Front Cover E. Wait for the lock to indicate unlocked V Reagent Compartment V Reagent Probe 1 V Reagent Probe 2 Connected to PCC V Reagent Probe 3 C V IM Acid V IM Base Patient Sample Processing V IM Cleaner D V IM Wash Front Cover V Special Reagent Water Back Cover V Cuvette Loader V Liquid Waste V Tin Trav Waste ? PEP connect LabManager Template Effective Date: 27-MAY-2019 | HOOD05162003010284 1 Effective Date: 25-Jan-2021| HOOD05162003159180 Unrestricted © Siemens Healthineers, 2021 Atellica® IM Analyzer SIEMENS Healthineers Clearing the IM Cuvette Loading Area Job Aid A 2. Inspect the IM cuvette loading area for jams: A. Inspect the cuvette bin and clear any jams. Remove cuvettes, if appropriate B B. Inspect the conveyor to clear jams. Remove cuvettes, if appropriate C. Inspect the orientation chute and clear any jams. Remove cuvettes, if appropriate C 3. Close the cover and select resume to bring the analyzer back to the Ready state Template Effective Date: 27-MAY-2019 | HOOD05162003010284 2 Effective Date: 25-Jan-2021| HOOD05162003159180 Unrestricted © Siemens Healthineers, 2021 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2021 Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 3 siemens.com/healthineers Effective Date: 25-Jan-2021| HOOD05162003159180 Unrestricted © Siemens Healthineers, 2021