Configuring Automatic Calibration Ordering on the Atellica® Solution Job Aid Version 1.25 and Below
This job aid provides instructions on how to configure automatic calibration ordering on the Atellica® Solution.
Atellica® Solution SIEMENS Configuring Automatic Calibration Ordering Job Aid Healthineers 1. Select Setup on the Command Bar 2. Select the Test Definition tab SIEMENS Atellico Solution Version 18.104.22.16874008 P-P1-03 K 1 3. Select either IM Test Definition or CH Test Definition Samples riventor Calibration oc Patient Orders Workdert Mantersince Setup 2 Test Definition Settings Accounts 4. Select the assay D Assay List Fiter Ranges 5 Select the Calibration tab CH Test Definition Definition Calculation 5. 4 Reflex Group Setup Measuring Interval Verification 3 IM Tast Definition BAM (8053) Generel Acceptance (ªAcceptance Calibration interval Repeat Conditions 6. Select the Automatically Order Calibrations checkbox Alc_3 (705) Automaticaly Order Calibrations 6 Ratios, Panels, and Off- System AAG (1109) Let Calibration Expiration [Daya) Lot Calibration Interval (Days) Restore Delsuts AAT (1110) 7. Select the criteria to trigger the system to Test Display Order Pack Calibration Lapiration (Hours) Acet [102) Pack Calibration Interval (Days) automatically order a calibration: Open Channel New Lot Alb (1106) 7 Calibrator Onboard Stability (Hours) Sort Test Definition NbP (148) IFU Changed Refrigerated 72 • Lot Calibration Expiration – set the number of days ALP_2e (1108) Perform QC with Calibration ALT (1105) Unrefrigerated 168 prior to the expiration of a lot calibration ALTPLC (4035) Ammm (116) • Pack Calibration Expiration – set the number of hours prior to the expiration of a pack calibration • New Lot • IFU Changed 1 Template Effective Date: 05/27/2019 | HOOD05162003010284 IM Effective Date: 8/23/2019 | HOOD05162003031090 Hardware Overview Unrestricted © Siemens Healthcare GmbH, 2019 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability may vary from country to country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2019 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 2 Effective Date: 8/23/2019 | HOOD05162003031090 Unrestricted © Siemens Healthcare GmbH, 2019