Controlled Aliasing in Parallel Imaging Results in Higher Acceleration (CAIPIRINHA)
Capering yeah stands for controlled aliasing in parallel imaging results in higher acceleration. It is an improved parallel imaging technique that makes optimal use of coil sensitivity profile information by adapting the case base undersampling. Compared to conventional parallel imaging caper, enia uses a case base undersampling, which is adapted to the coil sensitivity profiles. In fact, with type arenia, the acquired points in K space are shifted from one another by applying additional offsets in the phase encoding direction. An algorithm then reconstructs the missing gaps from the case base by combining information from the acquired points. Despite higher computational complexity, Kipe Arena reduces aliasing artifacts and noise and preserves higher signal to noise ratio than conventional P80 acceleration. Type ARENIA enables fast 3D, high resolution isotropic scans in different body regions and allow shorter breath holds without affecting image resolution coverage or contrast.
acquired points,case base,noise,aliasing,higher,enia,sampling,fact,parallel,operator manual,siemens healthineers,space,controlled,optional,hereinafter,collectively,country,training,caipirinha,material,information,acceleration,note,software,hardware,part,head,knee,present,functionalities
CAIPIRINHA Fully encoded image CAIPIRINHA image @ Siemens Healthcare GmbH, 2019 Controlled Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This Imiaging Aliasing material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily in reflect the latest version of the software and hardware available at the time of the training. The Operator Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Parallel Note: Some functions shown in this material are optional and might not be part of your system. The CAIPIRINHA Head Abdomen Knee information in this material contains general technical descriptions of specifications and options as Head well as standard and optional features that do not always have to be present in individual cases. Certain products, product related claims or functionalities described in the material (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. Due to Results Knee regulatory requirements, the future availability of said Functionalities in any specific country is not exe guaranteed. Please contact your local Siemens Healthineers sales representative for the most current in information. Hip The reproduction, transmission or distribution of this training or its contents is not permitted without Ankle/F00t express written authority. Offenders will be liable for damages. Higher All names and data of patients, parameters and configuration dependent designations are fictional Coil element 1 and examples only. No acceleration Acceleration All rights, including rights created by patent grant or registration of a utility model or design, are No acceleration Acceleration tl space Caipi accelerated (PAT=4) reserved. Coil element 1 TA = 08:32 min 02:14 min