Creating a SaveLog - Numaris X
This Job Aid provides workflow instructions on how to export create a SaveLog on Numaris X software to capture data from the scanner to be used later to evaluate or assess the MR systems functionality by the Siemens-Healthineers Customer Care Center (CCC).
Creating a SaveLog – Numaris X Creating a SaveLog The Customer Care Center (CCC) or your Customer Service Engineer (CSE) may ask you to create a SaveLog on your MR system. A savelog is performed to capture data from the scanner that can be used later to evaluate or assess the MR system’s functionality. Hints Perform the SaveLog at the time of error or incident, prior to restarting the system. Avoid performing any other operations on your scanner when performing a Savelog. 1. Home screen, upper right-hand corner, hover over Cogwheel Icon Select “SaveLog” Meduser ? Configuration Panel Administration Portal Remote Access Remote Assistance SaveLog 2. Create SaveLog window Your Name – Enter Facility name Describe the error or your observation – enter Description of error/observation Attach Screenshots – checkbox to include screenshots into the SaveLog file Select OK Create SaveLog Your Name Describe the error or your observation Screenshots Screenshots may contain patient data. S Screenshots help our service analyzing problems faster and more efficient. Screenshots will only be used for this purpose and will be handled confidentially Attach Screenshots OK Cancel Unrestricted Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 SIEMENS Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Job Aid Effective Date: 04/21/2020| HOOD05162003080107 Healthineers Job Aid | Creating a SaveLog-Numaris X Effective Date: 04/21/2020 | HOOD05162003080107 3. Create SaveLog dialogue box appears (DO NOT close this window) a. SaveLog process – takes approximately 3-5 minutes to complete C:\Program Files\Siemens\Numaris\bin\syngo.MR.SaveLogDump.exe - X Log: C:\ProgramData\Siemens\Numaris\log\syngo.MR.SaveLogDump.exe\syngo . MR.SaveLo gDump.exe_2020-03-24T17_52_32.5057345+01_00.txt 20200324-165232 ==================================== ========== 20200324-165232 Command line: "C:\Program Files\Siemens\Numaris\bin\syngo. MR.Sav e LogDump.exe" /pid 7328 /full 20200324-165232 Username=NT AUTHORITY\SYSTEM, Admin=True, Culture=en-US, UICultu re=en-US 20200324-165232 PID = 7328 20200324-165232 Name = syngo . MR.Protocol. Host.Ser 20200324-165232 Creating dump: C:\TEMP\caea7444\ Create SaveLog X Creating SaveLog .. Do not scan a patient while SaveLog is running! System Che 4. Created SaveLog is automatically stored into an Offline File (e.g., C:/store/SaveLog) Select OK – closes window Create SaveLog × V SaveLog created successfully The created SaveLog was saved to C:\store\SaveLog The created process dumps were saved to C:\store\SaveLog OK Unrestricted 0 Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 SIEMENS Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Job Aid Effective Date: 04/21/2020 | HOOD05162003080107 Healthineers Page 2 of 3 Job Aid | Creating a SaveLog-Numaris X Effective Date: 04/21/2020 | HOOD05162003080107 Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH 2020 Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84-0 siemens-healthineers.com Unrestricted 0 Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 SIEMENS Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Job Aid Effective Date: 04/21/2020 | HOOD05162003080107 Healthineers Page 3 of 3