Replacing LOCI Vacuum Cup Video
This video provides instructions on how to replace the LOCI® vacuum cup on the Dimension® EXL™ 200 System.
With the system in standby, enter the low side diagnostic screen by pressing F7 diagnostics. F1 electromechanical F2 sample system. After 7 loci module. Press F6 inspect to rotate the loci, arm out above the HM Wheel. Pull down on the vacuum Cup to remove it. Press the new vacuum cop on to the arm in reverse fashion, making sure it is attached firmly to the underside of the arm on the touch screen. Exit to the main operating screen by pressing the home key. The next step is to perform loci. Arm alignments refer to the operator's guide for specific details on this process. Finally, document this replacement on the instrument log sheet.
LOGS F2: PLUS Wheel F3: TEST RESULTS F3: ALIGNMENTS F3: Samp Wheel Run F2: Sampler F2: SAMPLE SYSTEM SAMPLE SYSTEM INSPECT F6: INSPECT LOCI Module F7: DIAGNOSTICS F7: KINETICS Fl: REAGENT ELECTRO/MECH Home F6: WATER WORKS F5: PHOTOMETER .11