Replacing HM Pump Heads Video
This video provides instructions on how to replace HM pump heads on the Dimension® Systems.
With the system and standby access, the HM Pump prime screen from the operating menu by pressing F for system prep. F7 pump prime. F6, HM. Raise the IMT and reagent lids. The aspirate and dispensed pump heads are located on the wash station frame. Using your fingers, squeeze the two tabs on the top and bottom of the pump head and remove from the pump base. Insert the shipping plug or T plug that came with the pump head completely into the front of the cassette. Slide the pump head on to the shaft of the motor, allowing the shipping plug to fall as the new pump head is installed, snap the pump head into place. Disconnect the tubing from the used pump at at its connectors and attached to the new pump head. Discard the used pump at into a biohazard waste container. Prime fluids by entering 3 into the cycle count field. And pressing F1 HM wash pump. Next, perform a system check. Refer to the operators guide for specifics on this procedure. Document this replacement on the monthly maintenance log sheet.
SN3W31S SN3B31S WATER PUMP F2: SAMPLER F3: SAM F3: RE, F3: REAGENT 1 F3: WIT F3: IMT 1M T F4: SYS I F4: RE SYSI NT PREP IPLER SIEMENS F4: SYSTEM C"ECK F4: SYS I SYS T RESULTS L PI-JMp L PUMp PUMP F 7: RE F7: PUMP PRIME F8: DAI 'UNTERS F8: DAILY MAINT. F8: SEGMENT STA SNOSTICS MP PRIME SIEMENS SELECT