Replacing Flex Insert R2 1500/3000T Video
This video provides instructions on how to replace R2 flex inserts on the Dimension Vista® System.
To begin cleaning the reagent arm to flex, insert, release the center interlock and lift the center lid. For easier access to the insert, remove the cuvet wash shield. Lift the reagent arm to flex insert out of the instrument to remove it for cleaning. Hold the insert under running water to clean it. And dry it thoroughly with a paper towel or gauze. Reinsert the reagent arm to flex insert, being careful to position in place. Tighten the thumbscrew to position properly. Reinstall the Cuvet wash shield. Now that you have finished cleaning the reagent arm, two Flex insert, close the center lid to access the software.
reagent arm,center lid,flex insert,flex,cleaning,insert,position,access,lift,remove,clean
Dimension [email protected] 50011000T (9 System Clean R2 Flex Insert