Cleaning Aliquot Waste Chute Video
This video provides instructions on how to clean the aliquot waste chute on the Dimension Vista® System.
The aliquot waste chute is located above Wastebin a cleaning. The aliquot waste chute requires no specific instrument status. Open the left and center left doors of the instrument. Remove waste bin A and locate the aliquot waste chute using a damp cloth. Thoroughly clean the shoot to remove patient sample residue. Replace Wastebin A and close all instrument doors. Record this procedure in the maintenance log.
SIEMENS ENS Dimension [email protected] System Clean Aliquot Waste Chute seq.