Elara Max Charging the System
This SBS will cover the proper steps for charging the Elara Max System.
Charging the system Elara Max The following steps will assist you with charging the MOBILETT Elara Max. Set the key switch to position O (key switch variant) or switch off completely using the ON/OFF button on the display panel (numeric lock variant) when you are not using the MOBILETT Elara Max. Connect the power cable to the socket. Ensure that the cable is laid neatly on the floor and does not form any loops or traps. The unit is always charged when it is connected to the mains, irrespective of whether the system is currently switched on or not. To ensure that the system batteries have a long lifespan, make sure that they are fully charged to 100% at least once a week. Battery charge indicators The battery charge indicators can be found on the control and display panel: Indicator Function Mains operation LED: System is connected to mains power 6 battery charge LED bars: Showing the current charge state of the system. LEDs running from left to right show that the unit is currently charging up. + From 6 to 2 bars: Unit is operable without restriction. SIEMENS Healthineers siemens.com/healthineers Job Aid | Charging the system Effective Date: 04/05/2019| HOOD05162003004657 1 bar left: Only a few exposures left. ----- Connect system to mains power immediately. 0 bars left: Red warning LED lights up ------ No exposures possible Only some operating time left. Connect system to mains power immediately. ERROR 225: ERROR 225 Error 225 is shown on the control and display panel after further discharge. System is switched off completely. Detector in holder stops charging. All LEDs are off. Connect system to mains power immediately. Siemens Healthineers Services Page 2 of 3 © Siemens Healthcare GmbH, 2017 Job Aid | Charging the system Effective Date: 19/11/2018| HOOD05162002971847 Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH 2018 ………………………………………………………… Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 siemens.com/healthineers Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2018