Elara/Mira Max: Performing an Emergency or Downtime Exam
This SBS will walk the user through how to perform an exam on both the Mira
and/or Elara Max systems when the RIS worklist is not working.
Performing an Emergency or Downtime Exam Mira or Elara Max If RIS worklist is down and you are unable to refresh your worklist or locate your patient you may use the following work-flow: 1. While in “Patient” sub-tab, select the icon to manually Patient Settings Service register patient: + FO RIS RIS 2. Enter any requested bold fields. • When patient information is not available (emergency patient), bypass bold fields and select “Emergency” in lower left corner. i. Select OGP views or exams you wish to complete and select “Examination”. ii. Perform Study 3. Once you have performed the exam, you may modify this study. • To do so, you will return to the main “Patient” tab and select the following icon from the “Patient” sub-tab. 4. Make any necessary changes to patient information: • Name, Patient ID#, DOB, Gender, or any other bold fields. • If an Accession number is now available, you may also enter it at this time. • When you have corrected any missing demographics you can return to patient exam and send images through the “Documentation” tab. SIEMENS siemens.com/healthineers Healthineers Job Aid | Performing Emergency Exam Effective Date: 02/27/2019 | Quality HOOD05162002996465 Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH ………………………………………………………… Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 siemens.com/healthineers Siemens Healthineers Services Page 2 of 2