epoc® Enterprise Data Manager QA Tests Job Aid
This job aid provides the instructions on how to navigate the QA Tests page.
QA Tests Page Job Aid epoc® Enterprise Data Manager The QA tests page displays Quality Assurance test records. From the Menu banner, select QA tests to view this page. Individual QA test details are displayed when the User selects a test from the Main Table. Single QA test details are displayed below the test table as shown in the following figure. To access a single test record, double click on a single record listed in the Main Table. Blood tests QA tests + Reports + | Inventory - | Settings > QA tests (Cust Today (34) Select filter type Last 7 days Last 30 days Date/Time QA test type Operator Depar 27-Jan-16 14:50:01 CV1:183-1-B525 Quality control 1111 Def 27-Jan-16 12:31:56 CV1:183-1-B525 Quality control 1111 Def Unrestricted Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2019 SIEMENS Master Template Effective Date: 3/11/2019 | HOOD05162002998095 Healthineers Job Aid Effective Date: 06/13/2013 | HOOD05162003012015 Job Aid | EDM QA Tests Page Effective Date: 06/13/2013 | HOOD05162003012015 QA test records are arranged with tabs for: A. Results B. Equipment C. Test change log D. Test results are displayed under the Results tab Note: If eVADs are used, test results will include a Status column indicating Pass or Fail results. The eVAD version will also be indicated. Refer to Appendix E “User Guide: Enhanced QA Features for the epoc System” in the end of this manual for details about obtaining and using eVADs. 14-Jul-15 13:00:54 CV1:183-1-B525 Quality control 1111 Default 0059AAF Rdr3610 IQC Fail Not sent 13-Jul-15 16:12:12 L3:179-3-B429 Calibration verification 1111 Julia 0059AAF Rdr3610 OK Pass Not sent 13-Jul-15 15:58:25 L3:179-3-B429 Quality control 1111 Default 0059AAF Rdr3610 OK Pass Not sent 13-Jul-15 15:16:55 CV1:183-1-B428 Calibration verification 1111 Default 0059AAF Rdr3610 OK Fail Not sent 50 Page 1 of 1 1 - 34 of 34 Save as CSV file Save chan B Cancel changes Printable report Send to LIS A D Results Equipment Test change log C Test record details Test results Test date/time 13-Jul-15 16:12:12 Analyte Result Reference range Status Lot ID L3:179-3-B429 DH 7.625 7.536 - 7.656 Pass QA test type Calibration verification pCO2 20.7 mmHg 17.5 - 27.5 Pass Comments pO2 190.0 mmHg 154.5 - 209.1 Pass Na+ 163 mmol/L 158 - 172 Pass K+ 6.1 mmol/L 5.6 - 6.8 Pass < > Ca++ 0.63 mmol/L 0.58 - 0.72 Pass Operator name Glu 262 mg/dL 203 - 317 Pass Operator ID 1111 Lac 6.62 mmol/L 4.73 - 7.47 Pass Department Julia cHCO3- 21.5 mmol/L Test Status OK cTCO2 22.2 mmol/L Fluid lot L3:179-3-B429 BE(ecf) 0.4 mmol/L Fluid ref 179.003.010 CS02 99.8 % Fluid type L3 A 124.4 mmHg Fluid expiry 31-Jul-15 A-a -65.6 mmHg eVAD version epoc15188.eVAD a/A 1.5 % QA status Pass Unrestricted Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2019 Page 2 of 3 Job Aid | EDM QA Tests Page Effective Date: 06/13/2013 | HOOD05162003012015 Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (herein- after collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH 2019 ………………………………………………………… Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 siemens-healthineers.com Unrestricted Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2019 Page 3 of 3