Exporting and Importing QC Definitions on the Atellica® Solution Video Version 1.25 and Below
This video provides instructions on how to export and import QC data on the Atellica® Solution. For optimal viewing, select the arrows on the right of the playbar after the video begins.
QC definitions can be exported as an XML file and then imported to another system. QC definitions can also be obtained from the supplier as an XML file and then import it to the system. To export a QC definition, first select QC at the command bar. Then select the QC definitions tab. The QC definition screen displays a list of the QC definitions currently on the system. Select the definitions to export and then select the export button. Select the folder location to place the exported QC definition XML file, and then select OK. A message displays indicating the definitions were successfully exported. After the export completes, the XML files can be obtained from the drive they were saved to. In this example, the files were saved to the exports folder in the operator data drive. To view the files in this folder, select the File Manager icon in the status bar and then navigate to the folder. When exporting multiple QC definitions, the system creates one export file for each control name. If multiple definitions have the same name, 1 file will be created for that name. Now let's discuss how to import a QC definition. To import an XML file, QC definition that was either obtained by exporting from an atellica solution or obtained from the supplier, select the import button on the QC definition screen. The import QC definition window displays. Select the browse button to locate the import file. In the select the browser window, pop up, select the browse button. Locate the import file, select the file, select open and then OK. The file name will display in the import QC definition window. If you want to only import QC definitions for tests that have been enabled on the system, select the checkbox. Select the import button. The import QC definition screen will display with the control name, Lot's expiration date and tests shown. The usage for each slot can be changed using the dropdown lists. Select the checkbox next to the test that you want to import the QC definitions. Select the save button to import the QC definitions. The newly imported QC definitions will display on the QC definition screen. This completes the overview of importing QC definitions.
qc definition,browse button,browser window,xml file,import button,import qc,import file,definitions,file,export,folder,import,files,system,select,obtained,supplier,saved,locate,display,liq tube,lot expiration,serum plasma,type,tests,serum,plasma,imen,control,usage
SIEMENS Solution Version 220.127.116.1146003 Healthineers Heotthineers Patient O Patent O File Explorer Create QC Orders OC QC QC Stat stics patient QC Flagging Patient ac Flagging QC QC Definitions Rack ac Definitions ac Panels QC panels panels Import QC Definition • with • Y status Control Name IA Plus Liq Tube Name IA Plus Liq Tube set 40930T + Contrd Name Ccntrol Name Name Level 3 Level Control 10 Control control Contre Lot Expirathn Onb»rd Usage Control 10 control Lot Expiration Lot Expiratdln usage Type Checksum specimen Specimen Sc«imen Onboard perform QC with Ccntml Name Ccntrol Name A Cmtrol N n e Ccntrol Name Control 10 Lol Lot Sp«imen Type Sc«imen Sp*2imen Type Sp«imen Date Last used Date I_æt used primary Primary 107/31/2040 042840931 T Cardiac Ma rkersT Cardac Ma rkersT Car"ac Ma rkersT 43023661 T 236611 04,'2SQ018 Select Tests Tests 10/31/2040 Primary 107/31/2040 Ma 07/31/2018 07/31/2019 Serum , Plasma Serum Plasma 07/371/2019 ModL& Ty•pe ModL& Type 107/31/2040 107/31/2019 04/27/2018 107/31/20„40 010/31/2040 Cardiac Ma rkersT MarkersT Pnimary primary 107/371/2019 04/2712018 07/271/2019 04/27/2018 Select the browser Window. Open Stability Stability St&lity Serum , Plasma Test Name usage Control Lot Control Level Units SO Type Low Select the browser Window. Alc_3 Import QC Import QC Definltlon Chem control 26430 42766791 T 101/31/'2019 07/31 '2018 ALP_2c X CCntroI Name X Ccntrol X Test Name Level Control ID Lol Lot Lot Expiration Lot Expiratdl onboard Replicates System Test Options 40932T 40933T CM PLUS 1 CM PLUS Disk Location 40933T 40932T CM 2 cm_2 CEA 3 Level CM 3 01/31,'2019 primary Primary 40931 T 40931 r 40932T 40933T D_HDL D-HDL 40932T 40933T Serum Plasma control Primary primary Prim aty Lol Serum Plasma 107/371/'2018 07/371/'2018 Serum , Plasma 04/2712018 0m 312018 0m 32018 Serum , Plasma 10032017 042940931 T HAVM pos 107/371/2019 04/27/2018 07/371/2019 Serum Plasma 10032017 10/032017 107/0371/2019 U0/032017 U0032017 Primary 6315471 6215472 07/271/2019 107/031/2019 10031/2017 05/29/2018 IA PLUS 1 Serum Plasma Serum , Plasma Serum ,Plasma 03/29201 g Save QC OC Import Import QC Add Add OC LabManager Tuesday, November 27, 2018 10:01 AM