This Step by Step will cover the process for placing or removing the footboard
of the fluoro table.
Footboard Removal/Replacement XPRF 1. To remove the footboard, position yourself at the back of the table. Luminos Agile Max 2. With both hands squeeze the right handgrip. SIEMENS siemens.com/healthineers Healthineers Job Aid | Footboard Removal/Replacment Effective Date: 11/29/2018 | Quality HOOD05162002976799 3. Lift the footboard up and away from the right accessory rail, rotating it to free it from the left accessory rail. Remove the footboard. Luminos Agile Max 4. To place the footboard back on the table, hook the left side onto the left accessory rail first. Siemens Healthineers Services Page 2 of 4 © Siemens Healthcare GmbH, 2017 Job Aid | Footboard Removal/Replacment Effective Date: 11/29/2018 | Quality HOOD05162002976799 5. Squeeze the right handgrip and lower the footboard until it is flush with the tabletop. Then slide the footboard to lock into position. Luminos Agile Max Note: Verify the footboard is locked into position. o Red indicates the footboard is not locked into place. o Green indicates the footboard is locked into place. Not locked in Locked in Siemens Healthineers Services Page 3 of 4 © Siemens Healthcare GmbH, 2017 Disclaimers*: Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH ………………………………………………………… Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 siemens.com/healthineers Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2017