Handling IM Analyzer Extended Downtime Job Aid Version 1.25 and Below
This job aid describes how to handle extended downtime on the Atellica® IM Analyzer.
Atellica® IM Analyzer SIEMENS Healthineers Handling IM Analyzer Extended Downtime Job Aid SIEMENS Healthineers Atellica Solution K A System Samples Inventory Calibration QC Patient Orders Worklist Preparing the IM Analyzer for Extended Downtime: Statu Logs Operator Diagnostics About System VMM P-P1-03 C-P1-06 P1-060 1. Ensure the IM Analyzer status is on: B A. On the Command bar, select System > Status Standby Standby Standby Standby B. Select the IM Analyzer Capabilities Alert Cate C. Ensure the analyzer is not paused or stopped Luminometer V Reagent Loader V Reagent Compartment V Reagent Probe 1 V Reagent Probe 2 Connected to PCC V Reagent Probe 3 V IM Acid (1) V IM Base V IM Cleaner V Patient Sample Processing V IM Wash Front Cover V Special Reagent Water Back Cover V Cuvette Loader V Liquid Waste V Tin Trav Waste ? PEP connect LabManager *This procedure is continued on the next slide Template Effective Date: 27-MAY-2019 | HOOD05162003010284 1 Effective Date: 25-Jan-2020| HOOD05162003159176 Unrestricted © Siemens Healthineers, 2021 Atellica® IM Analyzer SIEMENS Healthineers Handling IM Analyzer Extended Downtime Job Aid SIEMENS Healthineers Atellica Solution Version 220.127.116.1151003 2. Log off: System Samples Inventory Calibration Qc Patient Orders Worklist Maint On the Status bar, select the log off icon Status A. LOUR Operator Diagnostics Abou B. Select Sign Out ICH C. Select Yes Svosystems Descrigoon Gloshing the Reageitt Looder Clothing the Beabient Loader 3. Unloading onboard reagents is not Ceand the System Fluids Drawer Cleaning the Systenl Floas Thawte necessary as the analyzer continues to cool Cleaning Wither Probes B and mix reagents Probes Theposting Washer Probes Gearing and Inspecting the CH Probes and More Ironers Aulechess Ciparino ano inspecung Ine CH Probe Wash Stations Lock Sign Out Cancel Inspect and Clean A ? PEP connect LabManager 2 Effective Date: 25-Jan-2020| HOOD05162003159176 Unrestricted © Siemens Healthineers, 2021 Atellica® IM Analyzer SIEMENS Healthineers Handling IM Analyzer Extended Downtime Job Aid SIEMENS Resuming the IM Analyzer after Extended Downtime: Healthineers Atellica Solution Version 18.104.22.16851003 K 999 System Samples Inventory Calibration Qc Patient Orders Worklist Mainter 1. Sign in to the system: A. In the Status bar, select the log in icon B. At the Sign In window, enter the Operator ID and Password C. Select Sign In SIEMENS Healthineers' B Operator ID labmanager Password Forgot Password C ? Restart Shutdown. Sign In ? A PEP connect LabManager *This procedure is continued on the next slide 3 Effective Date: 25-Jan-2020| HOOD05162003159176 Unrestricted © Siemens Healthineers, 2021 Atellica® IM Analyzer SIEMENS Healthineers Handling IM Analyzer Extended Downtime Job Aid 3 SIEMENS Healthingers wer>> Atellica Solution Version " 14.0.2974006 P.P 2. Ensure the IM Analyzer status is on: A Symem Samples Inventory Calibration Oc Patient Orders Working Maintenance A. On the Command bar, select System > B Reagent Overview Reagent Loader Reagent Needs Supplies Overview Reagent Lot Comparison Cal-QC Storage Inventory Cal-QC Needs C Status 00104 CMOOTIO 00104 B. Select the IM Analyzer System Fluids Volume Status OBS Lot Expiration Lot Sequence Number C. Select the resume icon, if the analyzer is V IMAcid 10 OK 24d7h 2/6/2018 000008 0007 paused or stopped IM Base 11 OK 2447 h 2/6/2018 000008 0019 3. Check system fluids inventory and ensure IM Cleaner On reserve 5d 16h 5/8/2018 000008 0020 the IM Cleaner has not expired (see image) V IM Wash OK 26 d 16 h 9/23/2018 000028 1215 24 d 4h 9/23/2018 282482 1040 A. On the Command bar, select Inventory Replace System Fluids B. Select Supplies Overview Supplies Status Waste Status C. Select the IM Analyzer V Tos OK 2 Storage Locations Available 7 Tip Trays In Storage Cuvette and To Waste OK D. Check the status of the following: V Cuvettes OK To Tray Waste OK Open To Tray Waste Drawer i. System Fluids V Water OK Liquid Waste OK ii. Supplies iii. Waste E. If appropriate, replace system fluids or ? PEP LabManager Wednesday, June 06. 2018 1 :38 PM supplies, or empty the waste 4 Effective Date: 25-Jan-2020| HOOD05162003159176 Unrestricted © Siemens Healthineers, 2021 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2021 Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 5 siemens.com/healthineers Effective Date: 25-Jan-2020| HOOD05162003159176 Unrestricted © Siemens Healthineers, 2021