How to change cupping correction on Somaris 7 scanners
Job aid on steps to chane cupping correction on Somaris 7 scanners.
How To Change Cupping Correction on Somaris 7 SOMATOM Definition AS, Edge, Edge Plus, Flash, SOMATOM Drive, Force Cupping correction is an algorithm applied in the background that changes the image impression at the brain bone interface in head modes. This filter is applied automatically to the data when reconstructed. To review the different settings, follow the instructions to make a change and then restart the system. Create a new recon for the desired head exam and the new cupping correction filter will be applied. This can be done as many times as needed to obtain the desired impression. Note: Cupping correction only modifies the impression at the brain bone interface. If the overall impression needs modification, changing the reconstruction algorithm (kernel) should be com- pleted first. If scan parameters or patient positioning is not ideal, cupping correction may be lim- ited in regards to improvement. This setting should only be changed after discussion with the physicist or radiologist who is responsible for Image Quality. 1. From the Options menu, select Service, and then Local Service. Padent Applications Edit lasert View Setup Image Cotons System Help Version License Elle Browser .. Switch to External Simulator Hide Simulation Taskcard Maintenance Auto Expose Configuration Service Local Service Notes Remote Service Event Log. Start Problem Steps Recorder Anytime Software Update Remote Assistance Sere Log 2. The Local service login screen will launch. If there are characters in the service key field, backspace to remove. Once fields are empty, select OK. This will allow access in customer mode where only limited configuration changes can be made. SIEMENS Service Software SOMA TOM Definition AS Somaris/7 syngo CT VB20A Please enter Service Key OK Upload / Enter new Service Key So ecied by copy authorization. be greenwat win Siemess. case you have questions, please contact your local Siemens Service Organisation For tuithee information. services to ase contact your local Siemens Service Org SIEMENS Unrestricted © Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 Healthineers Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Job Aid Effective Date: 5/4/2017| HOOD05162002591627 Job Aid | How to Change Cupping Correction on SOMARIS 7 Effective Date: 11/01/2021| HOOD05162003230096 3. From the Service home page, select Configuration. SIEMENS Tome Monu Service Software - Siemens Expert SOMA TOM Definition AS Somaris/7 syngo CT VB20A Device Guard Enabled Configuration TuneUp Quality Assurance Filo & Image Tools Backup & Restore Control Test Tools Auto Report Open Service Tasks Service Cha DICOM Tests Maintenance SoftwareDistribution 4. Under Local Host, select Application. Cupping corrections options are listed to the right. 5. Select the desired setting. Note: To see general examples of what the impressions look like, select the Examples icon. 6. Select Save. A dialog box appears to confirm changes were saved. 7. Select OK to close the box and then select Finish. 8. To close the service home page, select the red X in top right corner. Configuration Service Key Expiry Date: Reports Home Help Logout System Options Application Specific Configuration -Local Host 4 View Direction Head · Cranial View Direction Body Caudal Site info TCPAP LAN Cranial Caudal Machine Application Topo Angle Lateral 00 0180 Technical Single Tube Maintenance Preferred ECG Color Coding DIEC/Europe OAHAUS Users MonitorType Display Patient Name on Gantry Panel - Security Settings Use RequestedProcedure as StudyDescription - - Service in SingleStudy mode 5 Mail FTP Cupping Correction Standard 01 02 03 04 Examples AutoTransfers Eventlog Dose Alert Password Allowed Characters Licensing Anytime Software U (6 Characters) [A-Z] [a-z].[0-9].[ ] Factory Enter Password: Escalatico Repeat Password: Helix -DICOM General Scan Protocol Password Allowed Characters Character Set (6 Characters) [A-Z].[a-z].[0-9].[ ] Structured Reports Enter Password: Offline Devices Network Nodes Repeat Password: Incoming Nodes Priet Devices HC Overview TeamPlay receiver LUT Files HIS/RIS Nodes -Import/Export 6 -External Devices V ready I Finish | Save SIEMENS Unrestricted 0 Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Healthineers Job Aid Effective Date: 11/01/2021| HOOD05162003230096 Page 2 of 5 Job Aid | How to Change Cupping Correction on SOMARIS 7 Effective Date: 11/01/2021| HOOD05162003230096 9. A system restart is needed to apply the changes fully. From the System menu, select End. The End Session dialog box appears. Patient Applications Edit Insert View Setup Image Options System Help ..................................................................................... End. Run 10. Select Restart System. After the system restarts, create a new recon to compare to previ- ous settings. End Session X og Off ... Shutdown System .. Restart System .. Restart Application. Standby. Cancel Help For questions or assistance, please contact CT apps support at the Customer Care Center 1-800- 888-7436. SIEMENS Unrestricted 0 Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Healthineers Job Aid Effective Date: 11/01/2021| HOOD05162003230096 Page 3 of 5 Job Aid | How to Change Cupping Correction on SOMARIS 7 Effective Date: 11/01/2021| HOOD05162003230096 Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH 2020 SIEMENS Unrestricted 0 Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Healthineers Job Aid Effective Date: 11/01/2021| HOOD05162003230096 Page 4 of 5 Job Aid | How to Change Cupping Correction on SOMARIS 7 Effective Date: 11/01/2021| HOOD05162003230096 Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84-0 siemens-healthineers.com SIEMENS Unrestricted 0 Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Healthineers Job Aid Effective Date: 11/01/2021| HOOD05162003230096 Page 5 of 5
- cupping correction
- image quality
- ct head