Installing Calibrator Definitions on the Atellica® Solution Video Version 1.25 and Below
This video provides instructions on how to install calibrator definitions on the Atellica® Solution. For optimal viewing, select the arrows on the right of the playbar after the video begins.
Calibrater definitions must be installed when a new lot of calibrater will be used to install a new calibrater definition on the system. The information can be scanned from the calibrater lot specific value sheet. This value sheet can be found in the box with the calibrating materials. To scan in a calibrater definition at the Command bar, select calibration. And then select the Calibrater definitions tab. Scan the barcodes from the calibrater lot specific value sheet using the barcode reader. The add calibrater definition screen will display when the barcode or bar codes have been successfully scanned, confirmed the scanned information that displays is correct. To enable the calibrater material for calibration, select the active checkbox. If the atellica solution is a sample handler configuration, there is an option to have the calibrating material stored in the Cal QC storage area. Select the store on board checkbox if you want the system to store the material. So like save to save the calibrater definition. The Calibrater definition now displays on the calibration definition screen. If the barcode reader is not available, calibrater definitions can be manually added by selecting the add new button on the calibration definition screen. In the add Calibrater definition screen, enter all of the information from the calibrater lot specific value sheet and then select the save button. To edit an existing calibrater definition on the Calibrater definition screen, select the calibrater definition and then select the edit button. On the add Calibrater definition screen, several items in the calibrater definition can be edited. Assays can be added, assays can be deleted, but level values can be changed. The expiration date can be corrected. The calibrater can be activated or deactivated, and the onboard storage option can be changed if the assay is able to be stored on board. To complete the procedure, select save to save the edited Calibrater definition.
calibrater definition,barcode reader,calibrater material,select save,calibrater,scanned,barcode,button,definitions,edit,added,save,select,information,calibration,expiration date,qc area,onboard storage,amp 1k,primewl —,cal code,calibrator mate,material id,assay type,add assay,calibrator material,calibrator,material,assay,board
SIEMENS SIEMENS . Atellicae Solution Version 184.108.40.20651003 Solution Version 1.1402951003 Atellicae Solution Version 1.1402951003 HeoLthineers• Healthineers SIEMENS SIEMENS . Add Calibrator Definition Healthineers Assay Type Material Name Material ID Material ID Expiration Date Calibrator Mate Calibrator Material Calibrator Material Calibrator Mate Add Calibrator Definition Store Onboard Atellica- 1M Tox cal C1022 7290999 Active Add Calibrator Definition Definitions calibration Measunng Interval Verification Material Measuning Interval Verification Material Material ID Matenial Name Lot Calibrator Mate rial Expiration Date Calibrator Mate Checksum Assay Type Material Name Material ID (O Calibrator Mate rial (O Calibrator Mate rial Calibrator Mate Calibrator Mate rial CAL •e/ Active •V Active C 1M Active (O Measuring Interval Verification Material Certain assay calibrators are not compatible witn onboard storage in the Cal-QC area. Measunng Interval Verification Material Measunng Interval Verifrcation Material (D Measuring Interval Verification Material Measuring Interval Verification Material 2017-12-27 Material Name Material ID Material ID Material 10 Expiration Date Assays Assays Assigned 5/1/2019 11:59 PM Alc_3, Ian assay calibrator that is not compatible with onboard storage in the Cal-QC area. Units Alc 3 siemens.com/document.library siemens.com/document-library Select Tests ASS AST Select CAL LOT Assay Units aHBcM Alc 3 Tnl_UL ng/mL CA125 CA19 9 Active ALTPLc ALTPLc CPLRIOO Amp 1K Dig EHV I Il I ' I I I I I I I I I I I I Il I Il I I I I I I Il : I I I I I I I Il ASO_2c ALP_2c A I ASO_2 ASO_2c ALTPLc AST ALTPLc d2Øb2S2B CHECKSUM primeAB0_A PrimeHP20_A PrimeHP20_A PrimeRP20_A primeWl —A Amp 1K Amp500 Note; To save the calibrator. add at least one assay PrimeHP20_A PrimneRP1_A Bnz200 PrirneRP3 APO Al APO B PrirneRP3 Bnz200 PrimeWI PrimeHP20_A PrimeWI0_A primeAB_A PrimeWI0 PrimneWI3 Bupn AST ASS UNITS C 208201911:59 PM primeW1 _A primeWl —A RgtA2 RgtB2 RP3VWVash RP3vwvash RP2VWVash Bupn Tnl_UL Add Assay by Cal Code Save Tnl_UL Add Assay by Cal Code LabManager Thursday. May 10.2018 8:30 AN Thursday. May 10, 2018 8:30 AM Thursday. May 2018 8:30 AM Thursday, May 2018 8:30