Manually Removing CH Reagent Packs from the Gripper Video Version 1.25 and Below
This video will explain how to manually remove CH reagent packs from the gripper on the Atellica CH Analyzer.
To remove a Ch reagent pack from the gripper on the atellica Ch analyzer, the analyzer must be stopped to stop the analyzer. First select system on the command bar. Next, select status. Then select the Atellica chemistry analyzer icon. If the analyzer is not already stopped, select the pause button to allow the system to complete any testing currently in process. The system will pause and then go to start state. Finally, select the button to unlock the front or back cover according to the gripper location. Open the cover. Once the system is stopped. You must open the cover within 30 seconds of unlocking, otherwise the cover will relock. To release the pack from the gripper, first grasp the bottom of the pack with one hand. Next using the other hand and a small flat edge screwdriver, turn the red screw on top of the gripper counterclockwise. Remove the pack. Do not pull the pack out on an angle from the gripper or it can result in contamination of the area. When the red screws loose and the pack should fall downward, an out of the gripper. Finally, using the screwdriver, turn the red screw clockwise until the gripper is closed. To complete the procedure, close the cover. Lastly, select the resume button in the software. If the pack is undamaged, reload the pack onto the analyzer.
red screw,button,pause,select,gripper,stopped,analyzer,cover,system,remove,stop,complete,finally,unlock,open,pack,hereinafter collectively,regulatory requirements,operator manual,siemens healthineers,operator,manual,solution,local,availability,material,training,country,siemens,front
SIEMENS Atellica• Solution Solution Heotthineers Please note that the learning material is for training purposes only! Operator Diagnostics For the proper use Of the software, please always use the Operator Manual Or Instructions for Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. CH Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software System and hardware available at the time Of the training The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Manually Remoue CH Reagent Packs Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. Capabilities Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local SIEMENS Siemens Healthineers sales representative for the most current information. from the Gripper on the [email protected] The reproduction, transmission or distribution Of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data patients, parameters and configuration dependent designations are fictional and examples only. Heotthineers All rights, including rights created by patent grant or registration of a utility model or design, are reserved. connected to pcc CH Analyzer Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands ore the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. 11 Copyright O Siemens Healthcare GmbH 2020 Front C.cwer Global Siemens Healthineers Headquarters Back Covet Henkestr. 127 91052 Erlangen, Germany Phone: +49 913184 0 siemens.com/healthineers