Manually Removing Tubes from the Atellica Decapper Video Version 1.25 and Below
This video will explain how to manually remove tubes from the Decapper on the Atellica IM Solution.
To manually remove YouTube from the decapper tube grippers, first turn the pressure valve at the back of the system clockwise to shut off the pressure of the system. Next ensure the status of the Atellica magline transport is paused. Navigate to the Magline status screen by first selecting system, then select status. Select the Atellica magline transport icon. View the status of the magline. If the magline is not already in an error, pause state. Pause the module by selecting the pause button and wait until the magline completely stops. Lift the decapper cover off the system. Set it aside and wait for the module to stop. To remove the tube from the tube grippers, grasp the tube grippers and pull away from the sample tube. Then remove the tube from the system. Be sure to hold the tube as the grippers are open so the tube does not fall and cause a spill. Replace the cover on the decapper. And then complete the procedure by turning the pressure valve counterclockwise to restore the pressure. If appropriate, resume the decapper an atellica magline transport by navigating to the appropriate system status screen and selecting the resume button.
tube gripper,mag line,pressure valve,pause,grippers,select,decapper,tube,pressure,status,selecting,remove,system,turn,module,wait,cover,hereinafter collectively,regulatory requirements,operator manual,siemens healthineers,operator,manual,solution,local,availability,material,training,country,siemens
SIEMENS Atelicoe Solution Atelico Solution Heoltnineers Please note that the learning material is for training purposes only! Ор«Х« For the proper use Of the software, please always use the Operator Manual Or Instructions for Use (hereinafter collectively "Operator Manual") ee ee issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. System p.' -03 ФЗ 0CR.01 ОСНИ OCR-Ol с.Р1-06 Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time Of the training The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Manually Remouing TUbes Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. Capabilities Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local SIEMENS Siemens Healthineers sales representative for the most current information. from the [email protected] Decapper Transport The reproduction, transmission or distribution Of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. Cover All names and data patients, parameters and configuration dependent designations are fictional and examples only. Heoltnineers p.' -03 р-р1-ОЗ с.Р1-06 All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Tube Gripper connected to pcc рсс Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright O Siemens Healthcare GmbH 2020 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 913184 0 siemens.com/healthineers