Mira Max: Proper Shutdown Procedure
This SBS will guide the user through the proper steps to shutdown the Mira Max system.
Proper Shutdown Procedure Mira Max It is recommended to reboot the system once a day, or once a week at minimum. To do so, press the Off button in the lower right corner of the numeric keypad: KV mAs 1 12 13 D A 4 5 6 A 7 8 19 A/D + - + OFF By selecting the Off button via the key pad, complete shutdown of the Computer and System in sequence will occur. This may take some time as active background applications are completed. o Do not try to switch the system on again immediately after switch-off. o Wait until the imaging system is shut down (dark monitor). o Then, wait a few seconds before switching on again. (*see pg. 39 of Fluorospot Compact Mira Max Manual for further detail). SIEMENS siemens.com/healthineers Healthineers Job Aid | Proper Shutdown Effective Date: 02/27/2019 | Quality HOOD05162002996470 Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH ………………………………………………………… Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 siemens.com/healthineers Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2017