Monthly Cleaning of the Atellica® Decapper Video Version 1.25 and Below
This video provides instructions on how to perform the monthly maintenance task of cleaning the Atellica® Decapper.
The operator must inspect and clean the camper area monthly to ensure the components are free from dust and debris. The tube gripper pads. Drip tray tube guide. And inside of the cap, wastebin must be inspected and cleaned. To clean the decapper components, pause the atellica magline transport by first selecting system. Then select status. Select the Atellica magline transport icon. And then select the pause button, wait until the atellica magline transport completely stops. Then turn the pressure valve at the back of the system clockwise to shut off the pressure to the system. Lift the decapre cover off the system and wait for the module to stop. Grasp the tube grippers and push them together. And then clean the tube gripper pads using a lint free cloth or cotton swab dampened with 10% bleach solution. Then clean the drip tray also using a lint free cloth dampened with 10% bleach solution. Anne clean the tube guide using a lint free cloth dampened with 10% bleach solution. Inspect the tube gripper pads to ensure they are secure and not worn. If the tube gripper pads are positioned incorrectly or excessively worn, contact the local Technical Support provider to replace. Ensure the cap remover is positioned over the cap waste chute. To clean the inside of the cap, wastebin pulled the cap waste drawer away from the system. Then remove the biohazard bag and discard according to laboratory protocol. Manually remove any caps dropped in the chassis. Clean the inside of the cap waste bin using a lint free cloth dampened with 10% bleach solution. Insert a new biohazard bag into the cap waste bin. Talking the bag over the bin an under the flange to allow the caps to drop freely into the cap waste bin. The new biohazard bag must be fully opened an fit snugly in the bin. Push the drawer closed until it is firmly against the back wall to ensure that it is in the correct position. If the drawer is open for less than 10 seconds, the system does not reset the waste counter. Replace the cover on the decapper. And then turn the pressure valve at the back of the system counterclockwise to restore the pressure. If appropriate, resumed the decapper an the atellica magline transport by first selecting system. Then select status. Select either the Decapper or Atellica magline transport icon as appropriate and then select the resume button. Note that if the magline needs to be restarted, follow the standard procedure of clearing the magline prior to startup. Refer to the Atellica solution online help For more information. To complete the procedure mark cleaning the decapper is completed in the software by first selecting maintenance. Then select schedule. Select the decapper module in the dropdown menu. Select the cleaning the decapper activity. And finally select the mark as completed button on the right side of the screen.
waste bin,bleach solution,biohazard bag,pressure valve,selecting system,select status,pressure,decapper,back,cover,completed,select,system,inspect,clean,hereinafter collectively,regular basis,necessarily reßlect,latest version,main reference,operator manual,manual,operator,module,healthineers,material,training,country,software,siemens
A&Lllica' Solution Version 22.214.171.12434003 Atellicae Solution Version 126.96.36.19934003 Atellicae Solution Version 1.18g.1 .3234003 SIEMENS . SIEMENS Atellica'• Solution 1 'e 1 3234003 -03 11 Heolthineers Healthineers System C alüün Calibration C auxün C altxün CahYation Cahtxation Püient Orders Patient Orders P*ient Orders Pütient Orders Pünt Orders P*ient M ailtenarEe M antenarx:e Sento Seno Setto Inve•tory Inventory P&nt Orders Maintenance Please note that the learning material is for training purposes only! Operator Status Report Operator Diagnostics O perator Diægnostics About Data Restore Software Software pdates For the proper use of the software, please always use the Operator Manual or Instructions for Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. System VMM p.p1-03 OCR-Ol DCR-OI c-P1-06 C-P1-06 c-P1-06 I-P1-06 C-P1-06 p 1-06 PI -06 Module: D-PP-02 Frequency: All Time: All Performed By All Any material used in this training will not be updated on a regular basis and does not necessarily reßlect the Latest version of the software Any material used in this training will not be updated on a regular basis and does not necessarily reßlect the latest version of the software and hardware available at the time of the training. D-PP-02 Module Module Activity Module Activity Scheduled Sun-24 Tue-26 Wed-27 Thue-28- Thue-28 Thu-28 Sat-02 The Operator's Manual shall be used as your main reference, in particular for relevant safety information Like warnings and cautions. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Frequency D-PP-02 D-P1-06 Autocheck 2/28/2019 2.128/2019 Standby Standby Standby Standby Note: Some functions shown in this material are optional and might not be part of your system. All Time D.PP.02 Clearang the (Monthly) 7:00 Not Perform... Not Perform Diagram Alerts Time Tune Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be commercially available in your All Performed By Performed Capabilities Preventive Maintenance Preventive Maintenar%e 1/1/2020 0:00 0:00 country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact SIEMENS SIEMENS . your local Siemens Healthineers sales representative for the most current information. Search Text Waste Bin waste Bin SH Autocheck Atellica Magline Transport The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be Cover Scheduled by Operator liable for damages. Healthineers Schedulemtomatically SC heduled Automatically Assembly All names and data of patients, parameters and configuration dependent designations are fictional and examples only. 1-03 -03 D-pp-04 c-P1-06 C-P1-06 Scheduled by Semce Scheduled by Service Scheduled by Serv•ce Connected to PCC All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Failed VMM Atellica Magline Transport Autocheck Overdue Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affliates. @ 11 All other trademarks and brands are the property of their respective owners. ox Inprocess 11 Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Completed X Not Performed Copyright O Siemens Healthcare GmbH 2019 With Comments Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany 1M Autocheck Phone: +49 9131 84 0 siemens.com/healthineers Add Activiy__ Add Activiy Set Warning__ Set Warning._ Set Warning _ Wednesday, December 12, 2018 12:46 Wednesday, December 12, 2018 1246 PEP LabManager Wednesday, 12, 2018 1246 Wednesday, 27, 2019 241 PM Wednesday, 27, 2019 242 PM Monday, 01, 2019 1239142 PM Monday, April 01, 2019 123942 PM April 01, 2019 123942 PM Aprily, 01, 2019 123942 PM