Omnia MAX Troubleshooting Guide
This SBS will guide users through common system circumstances that may
occur and how to resolve them.
Omnia MAX: Trouble Shooting Guide XP Error Code Notifications on Exam Room Monitor A If desired table movements or general imaging functions do not occur as expected during system operation look at the following to resolve. • Check the table’s left-side Montior A in the exam room to see if there is a descriptive Error Code notification will appear in the bottom-right corner of Monitor A as well as at the control station monitor. o There will be a descriptive reason for why the operation is not functioning, upon which corrective actions may be taken. T A small pop-up Error Code notification would appear in the bottom-right corner of the table's left-side Monitor A Omnia MAX system will not turn on or boot up, Or the table will not move after boot-up • Check the 3 red Emergency Stop Buttons (ESB) are not depressed inward – especially the Emergency Stop button on the open-side of the bed below the table top. o To deactivate any Emergency Stop Button, turn the depressed button one- quarter turn to the right or clockwise. Note: DO NOT use any Emergency Stop button as the system’s normal “ON & OFF” function! Only use this function for Patient Emergencies! • Make sure the on-site main switch (or circuit breaker) providing the line power for the system is switched on. May be found on the wall in the control room to the left of the system Tech station. • Press the ON button again on the control panel in the control room. SIEMENS siemens.com/healthineers Healthineers . Job Aid | Omnia Trouble Shooting Guide Effective Date: 10/01/2018 | Quality HOOD05162002955153 Example of Emergency Stop Buttons ESB ESB Locations of Emergency Stop buttons on cysto table ESB Location of the Emergency Stop button at tech station Uroskop will not X-ray or Fluoro • Verify the X-ray tube is fully extended out-ward over the bed. o The imaging system will not emit radiation if the X-ray tube assembly is not fully extended. o Depress and hold the Tube In/Out (Park) button on any hand control to extend the X-ray tube for imaging. o If the LED in the Tube In/Out button is off, then the tube assembly is in the working position and is ready to image. QP X-Ray Tube In / Out button on hand control There is no endoscopic or ultrasound imaging on the table’s right-side Monitor B • Depress the Endo Cam (Reference) button on any hand control until the active imaging appears. • The Physician can depress the top-centeral blue golf-ball button labeled Video on the foot switch until the active imaging appears. REF 0.00 Endo Cam (Reference) button on hand control Siemens Healthineers Services Page 2 of 4 Job Aid | Omnia Trouble Shooting Guide Effective Date: 10/01/2018 | Quality HOOD05162002955153 The table will not tilt head-down or foot-down • The “forearm” boom supporting the table’s dual-monitor frame is too close to the X-ray tube and may cause a collision upon table tilt. • Move the monitor frame just an inch or two further away from the X-ray tube. This arm is past 88 degrees (too close to Xray Tube) and causes an interlock of no table movement Siemens Healthineers Services Page 3 of 4 Job Aid | Omnia Trouble Shooting Guide Effective Date: 10/01/2018 | Quality HOOD05162002955153 Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH ………………………………………………………… Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 siemens.com/healthineers Siemens Healthineers Services Page 4 of 4