Patient QC Flagging on the Atellica® Solution Job Aid Version 1.25 and Below
This job aid provides instructions on how to use the patient QC flagging feature on the Atellica® Solution.
Atellica® Solution SIEMENS Patient QC Flagging Job Aid Healthineers The Patient QC Flagging feature allows you to configure the system to Atellico" Solution Version 22.214.171.12463002 SP00107 flag patient results based on QC error rule violations. E System Sarroles Mitverbory Calexation Patient Orders Woddel Mantervince Setup Create DC Ondars Patient QC Fisaging DC Schedules • To enable Patient QC Flagging, navigate to QC > Patient QC Flagging. Select a test and then select Edit/View. Select Automatic or Manual, Whole Boo and then select Save MO - • If a QC result triggers an alert that has been configured in the QC Patient OC Flagging : AAG x Statistics > Setup > QC Rules screen, the system will start flagging patient results for that test Semm ➢ On the Worklist Overview screen, the flag displays for ? Circel patient test results indicating that a QC violation error occurred for at least one level S ALIFE • Once the flagging has initiated for a test, the flagging is cleared by ALTPLE one of these methods, depending on the configuration settings: 100000000000000000000 Automatic: Patient QC Flagging is reset automatically after the QC ? Labilitanager Thursday, May 31, 2018 2 41 PM & levels that previously failed produce passing results OR Manual: Patient QC Flagging is reset when the operator selects the Clear Patient QC Flagging button on the Patient QC Flagging screen • When Patient QC Flagging is cleared, all patient results that had been flagged will stay flagged, new results will not be flagged 1 Template Effective Date: 27-MAY-2019 | HOOD05162003010284 Effective Date: 06-APR-2020 | HOOD05162003077519 Unrestricted © Siemens Healthcare GmbH, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability may vary from country to country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 2 Effective Date: 06-APR-2020 | HOOD05162003077519 Unrestricted © Siemens Healthcare GmbH, 2020