Performing Automated Atellica® Direct Load Autocheck Job Aid Version 1.25 and Below
This job aid provides instructions on how to perform the daily maintenance task of Automated Atellica® Direct Load Autocheck.
Atellica® Direct Load SIEMENS Performing Automated DL Autocheck Job Aid Healthineers Performing Automated DL Autocheck Activity SIEMENS Build 1.7.0 DL-P1-07 The DL Autocheck is an automated maintenance activity 1 System Samples Inventory Calibration CC Patient Orders Workist Maintenance Setup that tests the health of various DL components. 2 Schedule Report Log Diagnostics Data Restore Software Updates Copy Configuration Filter Module: All Frequency: All Time: Next 24 hours Performed By: All The system alerts the operator 15 minutes before the Modulo Al Module Activity Scheduled 23 Sat-29 Sun-30 Mon-31 Tue-01 Wed-02 Thu-03 4 Perform ... scheduled activity begins. After the system initiates the Frequency All C-P1-03 -P1-03 CH Weekly Maintenance 7/30/2017 4:00 V Mark as Completed DL Autocheck, the DL pauses all loaded racks until the Time Next 24 hours C-P1-03 -P1-03 Checking Lamp Coolant 7/30/2017 6:00 -- V Not Perform .. Performed By All C-P1-03 autocheck completes in approximately 3 minutes. C.P1.03 CH Daly Mantenance 7/31/2017 4:00 V Search Text Enter the activity name Procedure Details CP1-03 -P1-03 Inspecting Washer Probes : 7/31/2017 5:00 V V The task will run automatically daily but can be run Scheduled by Operator Edit/View .. Database Backup .:: 8/1/2017 0:30 V V Scheduled Automatically manually if needed: DL-P1-01 3 Of Scheduled by Service V DL-P1-01 Autocheck for DL 8/1/2017 1:00 Delete e CP1-03 IMT Daily Cleaning 8/1/2017 2:00 Failed V v 1. On the Command bar, select Maintenance >verdue Database: Archiving and Deleting 8/1/2017 3:00 OF Inprocess Database: Optimizing : 8/1/2017 5:30 V 2. Select the Schedule tab V Completed C-P1-03 P1-03 Replacing Source Lamp 9/1/2017 0:00 -- X Not Performed C-P1-03 3. Select Autocheck for DL With Comments C-P1-03 CH Replacing Dilution Cuvette Segments 9/1/2017 0:00 -- -- -- -P1-03 -P1-03 CH Replacing Reaction Cuvette Segments 9/1/2017 0:00 Miow Acinobes -- -- 4. Select Perform C-P1-03 C-P1-03 Replacing IMT Perstaltic Pump Tubing 1/1/2018 0:00 Add Activity ... Set Warning .. 5. Wait for autocheck to complete ? LabManager Monday, April 17, 2017 10:02 AM to HOOD05162002852946 Effective date: 5/21/2018 1 Unrestricted © Siemens Healthcare GmbH, 2018 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH 2018 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers HOOD05162002852946 Effective date: 5/21/2018 2 Unrestricted © Siemens Healthcare GmbH, 2018