Performing Automated Atellica Magline® Transport Autocheck Job Aid Version 1.25 and Below
This job aid provides instructions on how to perform the daily maintenance task of Automated Atellica Magline® Transport Autocheck.
Atellica® Sample Handler SIEMENS Performing Automated Atellica Magline® Transport Autocheck Healthineers Job Aid SIEMENS Build 1.7.0 DL-P1-07 The Atellica Magline Transport Autocheck is an automated 1 System Samples Inventory Calibration QC Patient Orders Worklist Maintenance Setup maintenance activity that tests the health of various 2 Schedule Report Log Diagnostics Data Restore Software Updates Copy Configuration Atellica Magline Transport components. Filter Module: All Frequency: All Time: Next 24 hours Performed By: All Module All Module Activity Scheduled Sun-28 Mon-29 Tue-30 Wed-31 Thu-01 Fri-02 Sat-03 4 Perform ... The system alerts the operator 15 minutes before the Frequency All CH GGCKKGSCI-CH IMT Daily Cleaning :: 05/31/2017 2:00 V V V Mark as Completed scheduled activity begins. After the system initiates the Time Next 24 hours CH GGCKKGSCI-CH CH Daily Maintenance : 05/31/2017 4:00 V V Not Perform .. Atellica Magline Transport Autocheck, the Atellica Magline Performed By All IM GGCKKGSCI-IM IM Autocheck 05/31/2017 3:30 V V V V Search Text Enter the activity name. SH Procedure Details GGCKKGSCI-SH Transport pauses transporting samples until the SH Autocheck :: 05/31/2017 1:00 V Scheduled by Operator VMM 3 Edit/View ... VMM Atellica Magline Transport Autocheck :!: 05/31/2017 1:00 V V autocheck completes in approximately 10 minutes. Scheduled Automatically Scheduled by Service Database Backup :: 06/01/2017:0:30 V V Delete IM The task will run automatically daily, but can be run GGCKKGSCI-IM Cleaning the Water Container 06/01/2017 1:00 V Failed Overdue GGCKKGSCI-IM Inspecting and Emptying the IM Water Trap : 06/01/2017 1:00 V V manually if needed: Inprocess Database: Optimizing .:: 06/01/2017 3:00 V IM 1. On the Command bar, select Maintenance Completed GGCKKGSCI-IM IM Weekly Maintenance :06/01/2017 5:00 V V x Not Performed Database: Archiving and Deleting .:: 06/01/2017 7:00 -- -- With Comments 2. Select the Schedule tab IM Www Activibos GGCKKGSCI-IM Cleaning the Exterior of the Aspirate Probes 06/01/2017 7:00 IM GGCKKGSCI-IM Cleaning the Reagent Probe Rinse Stations 06/01/2017 8:00 3. Select Atellica Magline Transport Autocheck Add Activity .. Set Warning. 4. Select Perform ? B) LabManager Monday, April 17, 2017 10:02 AM 5. Wait for autocheck to complete NOTE: Do not position pacemakers or other implanted medical devices closer than 7.5 cm (3 inches) from the Atellica Magline® Transport. Carriers on the Atellica Magline Transport generate strong magnetic fields that can interfere with cardiac medical implants. HOOD05162002822815 © Siemens AG 2013. All rights reserved. Effective Date: 3/12/2018 Page 1