Performing Automated IM Daily Maintenance on the Atellica® IM Analyzer Job Aid Version 1.25 and Below
This job aid provides instructions on how to perform the daily maintenance task of Automated IM Daily Maintenance for the Atellica® IM Analyzer.
Atellica® IM Analyzer SIEMENS Performing Automated IM Daily Maintenance Job Aid Healthineers IM Daily Maintenance contains automated maintenance SIEMENS . Healthineers Atellica Solution Version. 1.14,0.2983003 P-P1-03 activities. The system performs a series of cleaning and 3 4 System Samples Inventory Calibration Qc Patient Orders Worklist Maintenance Setup rinsing operations to decontaminate the analyzer at Schedule Report Log IMT Data Restore Software Updates Copy Configuration configured times or on demand. The operator can Filter Module: P1-060 Frequency: All Time: Next 24 hours Performed By: All configure the time this is performed. Module P1-060 6 Module Activity Scheduled Sun-10 Mon-11 Tue-12 Wed-13 Thu-14 Fri-15 Sat-16 Perform .. Frequency All P1-060 P1-06 Cleaning the IM Fan Filter . 6/1/2018 7:00 -- Mark as Completed To perform the Daily Maintenance activity manually: Time Next 24 hours 1-060 -P1-06 Cleaning the Exterior of the Aspirate Probes 6/1/2018 7:00 -- Not Perform. Performed By All P1-060 I-P1-06 Cleaning the Exterior of the Reagent Probes :6/7/2018 1:00 1. Ensure the analyzer is in ready or standby state Search Text Enter the activity name P1-060 Procedure Details 4-06 Inspecting and Emptying the IM Water Trap : 6/7/2018 1:00 -- Scheduled by Operator 5 Edit/View .. 2. Ensure cuvettes and IM Cleaner are loaded on the 66 Inspecting and Emptying the IM Dryer :6/11/2018 7:00 Scheduled Automatically P1-060 Scheduled by Service V 1-P1-06 IM Daily Maintenance :!!:: 6/13/2018 2:00 Delete V V system P1-060 I-P1-06 IM Autocheck .:: 6/13/2018 3:30 V V Failed P1-060 On the Command bar, select Maintenance Overdue -P1-06 IM Weekly Maintenance 3. A Confirmation 1-060 Inprocess 1-P1-06 Replacing the Aspirate Probe P1-060 4. Select the Schedule tab Completed -P1-06 Replacing the Aspirate Probe 200 X Not Performed >1-060 P1-06 Replacing the Aspirate Probe With Comments 5. Select IM Daily Maintenance P1-060 I.P1-06 View Achaties Replacing the Aspirate Probe Replacing the Reagent Probe ? Select Perform The maintenance activity IM Daily Maintenance requires 20 6. minutes. Perform this activity now? 7. Select Yes ? PEP. LabManager 8. Wait for maintenance to complete 7 Yes No Note: IM Daily Maintenance is not required when the system performs IM Weekly Cleaning. Note: If the cleaning procedure stops before it completes, the system logs an error and does not allow processing samples until it completes the cleaning procedure or completes a rinse procedure. 1 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 2 Effective Date: 20-DEC-2020 | HOOD05162003152571 Unrestricted © Siemens Healthineers, 2020