Performing CH Weekly Maintenance on the Atellica® CH Analyzer Video Version 1.25 and Below
This video provides instructions on how to perform the weekly maintenance task of CH Weekly Maintenance on the Atellica® CH Analyzer.
During the automated weekly maintenance activity for the Atellica Chemistry Analyzer, the system performs a series of cleaning and rinsing operations to decontaminate the analyzer. This activity will be run automatically on a weekly basis, but can be run on demand as needed. To manually perform chemistry weekly maintenance, first ensure the analyzer is in ready processing or standby state by selecting system in the command bar. Select status next, select the Ch module. Finally, review the status of the analyzer. To ensure the waterbath additive is on board, select inventory in the command bar. And then the reagent overview tab verify that the water bath additive is on board. Next, select the maintenance tab. Select schedule. Select the module icon from the dropdown menu. Selects weekly maintenance. Select perform. Then select yes, wait for the maintenance to complete. This completes the overview of performing Ch weekly maintenance.
command bar,select,maintenance,weekly,module,status,analyzer,activity,ensure,complete,reagent,front cover,reagent loader,atellicae solution,reagent overview,atellicae,onboard,inventory,automatically,overview,system,standby,cleaning,board,siemens,version 1,healthineers,reagents,calibration,ventory
SIEMENS Atellicae AtelLicae Solution Atellicae Solution Version 1. Atellicae Solution SIEMENS . Atellica Solution Version 1.1402975003 Atellica Solution Version 18.104.22.16875003 Solution Version 1. sp00107 Version 1. 14.0.2983003 14.0 2983003 140.2975003 0.29830 3 HeoLtnineers Healthineers SIEMENS . Healthineers I ry Reagent Overview Reagent Loeder Reagent Loader Reagent Loader Operator Diagnostics Reagent Needs Reagent Loader Overview S Overview Reagent Lot Comparison Reagent Lot Cal-QC Storage Needs SH 1M System c-P1-05 Module Activity Scheduled Reagents Reagent Details Reagent Needs Reagent Overview Onboard calibration Calibration cal Type QC Cleaning 2:00 200 Analyzer Reagents In ventory Inventory Sequence cal Eligibility Interval In Use 'n use Status Standby Standby TSH3UlL TSHWUL TSHWL 110 11 Not Calibration Confirmation Capabilities p 102 003416 Odi.Oh Odi.O'h 21d5h c.P1.os Weekly Odi.Oh 2 d h 00778 00688 00683 73d23h 13Od22h Lot and Pack this C.PI.OS 13d22h IOd22h Connected to FCC 21 doh 21dOh / 11391 111391 4 d Oh c-P1-05 42d22h 21dO3h 425d23h 225d23h 00368 00224 00692 00364 00688 00683 OdO.1h c-P1-05 125d13h 125 d 13 h 007152 00778 Front Cover 3d5h Back Cover Front Cover The Water Bath Additive (WBA) is loaded and unloaded onto the system the same as all other reagents. The With Comments WBA will automatically be kicked off ot the system when the Onboard Stability and/or expiration date passes. p 102 Group by reagent PEP rl PEP PE LabManager Wednesday. 17.2018 '.031AM Wednesday, 17, 2018 10:31 AM 2018 10:31 AM Tuesday, July 10.2018