Performing IMT Routine Clean on the Atellica® CH Analyzer Job Aid Version 1.25 and Below
This job aid provides instructions on how to perform the as needed maintenance task of IMT Routine Clean for the Atellica® CH Analyzer.
Atellica® CH Analyzer SIEMENS Performing IMT Routine Clean Job Aid Healthineers 1. At the Command bar, select Maintenance Healthineers Atellica Solution Version 220.127.116.1183003 SP00107 2. Select the IMT tab 1 System Samples Inventory Calibration Patient Orders Worklist Maintenance Setup 3. Select IMT Clean and Prime Schedule Report Log 2 IMT Data Restore Software Updates Copy Configuration CH 4. Select IMT Routine Clean CH 3 IMT Clean and Prime IMT Details IMT Messages IMT Status IMT Calibration completed 1. Select the appropriate IMT function. Prime IMT ➢ An IMT Calibration automatically begins when the IMT IMT Dilution Check MT Calibration 2. Select Prime fluids one at a time by selecting a fluid. 4 IMT Routine Clean Last Performed 06/20/2018 9:01:08 PM Routine Clean completes Next Scheduled 06/21/2018 1:01:08 AM Operation Status IMT Advanced Clean A-LYTE Multisensor 5. Wait for the calibration to complete Prime All Date Last Replaced 05/29/2018 1:41:45 PM Tests on Chip 0 Prime Standard A Routine Clean Prime Standard B Last Performed 06/20/2018 2:05:54 AM Advanced Clean Prime Salt Bridge Prime IMT Diluent Last Performed Not Performed Days to next clean Maximum Serum or Plasma samples processed 0 Limit Average Fluid Delta Not Available Minimum = 0.0 mv Sample Port Drain Time Not Performed Maximum = 1.7 seconds Cancel CD ? PEP )LabManager 9x Thursday, June 28, 2018 12:36:26 PM 1 Template Effective Date: 05/27/2019 | HOOD05162003010284 Effective Date: 24-SEP-2020 | HOOD05162003126734 Unrestricted © Siemens Healthcare GmbH, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 2 Template Effective Date: 05/27/2019 | HOOD05162003010284 Effective Date: 24-SEP-2020 | HOOD05162003126734 Unrestricted © Siemens Healthcare GmbH, 2020