PhaseRev Add-In- XA31
This Job Aid provides a short introduction to the PhaseRev Add-in and how it works.
PhaseRev Add-In Job Aid 1. What is the PhaseRev Add-In? • Add-ins are auxiliary software components. As a kind of plug-in assigned to a program step, they enhance it with specific features. • The PhaseRev (=reverse phase) Add-In copies a current step in the exam queue and flips the phase-encoding direction of it. • It can be attached to any protocol. • The same PhaseRev functionality is also directly built into the BOLD Add-In. 2. How can it support you? • DTI image may be affected by susceptibility-induced geometric distortions. Especially in a research context, third party post-processing tools (e.g. TOPUP • tool available as part of the FSL library*) offers the possibility to correct these geometric distortions based on additional image data acquired with opposed phase encoding direction compared to the affected data. With the PhaseRev Add-In you can automatically acquire additional data with • opposed phase encoding direction without manual user interaction to simplify the workflow and to ensure consistent data input for tools like FSL TOPUP*. *http://fsl.fmrib.ox.ac.uk/fsl/fslwiki/topup Unrestricted Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2021 SIEMENS Master Template Effective Date: 3/5/2020 | HOOD05162003074494 US Job Aid Effective Date: 4.12.2021 | HOOD05162003184902 Healthineers Job Aid Effective Date: 13-Feb-21 I HOOD05162003162958 Job Aid I PhaseRev Add-In Job Aid US Job Aid Effective Date: 4.12.2021| HOOD05162003184902 Effective Date: 13-Feb-21 I HOOD05162003162958 3. How does it work? 1. Open the PhaseRev Add-In, if you Dot Add-In Configurator want to use the PhaseRev Add-In without any further Add-Ins. Within the PhaseRev Add-In, the Guidance Parameters Enable PhaseRev functionality is automatically Queue GSP enabled. Series Label Auto Load AutoCoverage PhaseRev Close Please note: If you want to use the BOLD Add-In and the PhaseRev functionality in parallel, you need to use the BOLD Add-In. The PhaseRev functionality is disabled per default in the BOLD Add-In. To enable the PhaseRev functionality in the Add-In, put a checkmark in the Enable PhaseRev field. 2. Start a protocol in the exam queue e.g. gre protocol, that has the PhaseRev Add-In attached. The Add-In needs to be attached to the protocol in the Dot Cockpit as shown in the previous step. The phase encoding direction of the clinical protocol is e.g. A>>P. gre 00:14 Abdomen + PhaseRev gre gre_flipped 00:14 N - E 00:14 Continue Copy&co Go Cancel SIEMENS Unrestricted Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Healthineers Job Aid Effective Date: 13-Feb-21 I HOOD05162003162958 US Job Aid Effective Date: 4.12.2021 | HOOD05162003184902 Job Aid I PhaseRev Add-In Job Aid US Job Aid Effective Date: 4.12.2021| HOOD05162003184902 Effective Date: 13-Feb-21 I HOOD05162003162958 3. As soon as the scanning of the current protocol step is performed, another protocol step is generated by the PhaseRev Add-In with the flipped phase encoding direction P>>A. 4. The copied protocol step with the flipped phase-encoding direction is automatically created and inserted in the queue under the “original protocol name + flipped”. Abdomen ISO ℮ + 1 gre 2 gre_flipped 00:14 E 00:14 DI Go Cancel Delete Skcip Stop Pause Continue Copy&Go Please Note: Finish the original protocol step, to open the flipped protocol step. SIEMENS Unrestricted Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Healthineers Job Aid Effective Date: 13-Feb-21 I HOOD05162003162958 US Job Aid Effective Date: 4.12.2021 | HOOD05162003184902 Job Aid I PhaseRev Add-In Job Aid US Job Aid Effective Date: 4.12.2021| HOOD05162003184902 Effective Date: 13-Feb-21 I HOOD05162003162958 Please note that this learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH 2021 Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84-0 siemens-healthineers.com SIEMENS Healthineers